- Research article
- Open Access
Simultaneous determination of cetirizine, phenyl propanolamine and nimesulide using third derivative spectrophotometry and high performance liquid chromatography in pharmaceutical preparations
Chemistry Central Journal volume 11, Article number: 99 (2017)
The combination between cetirizine (CET), phenylpropanolamine (PPA) and nimesulide (NMS) under trade name Nemeriv Cp tablet is prescribed for nasal congestion, cold, sneezing, and allergy. Among all published methods for the three drugs; there is no reported method concerning estimation of CTZ, PPA and NMS simultaneously and this motivates us to develop new and simple methods for their assay in pure form and tablet preparations.
Two new methodologies were described for the simultaneous quantification of cetirizine (CTZ), PPA and NMS. Spectrophotometric procedures relies on measuring the amplitudes of the third derivative curves at 238 nm for CTZ, 218 nm for PPA and 305 nm for NMS. The calibration graphs were rectilinear over the ranges of 8–90 µg/mL for CTZ, 20–100 µg/mL for PPA and 20–200 µg/mL for NMS respectively. Regarding the HPLC method; monolithic column (100 mm × 4.6 mm i.d) was used for the separation. The used mobile phase composed of 0.1 M phosphate buffer and methanol in the ratio of 40:60, v/v at pH 7.0. The analysis was performed using UV detector at 215 nm. Calibration curves showed the linearity over concentration ranges of 5–40, 10–100 and 10–120 µg/mL for CTZ, PPA and NMS.
Application of the proposed methods to the laboratory prepared tablets was carried out successfully. The results were compared with those obtained from previously published methods and they were satisfactory.
Cetirizine (CTZ, Fig. 1a); is non-sedating antihistamine with long acting activity for treatment of urticarial and rhinitis . It is ([2-[4-[(4-chlorophenyl) phenylmethyl]-1-piperazinyl] ethoxy] acetic acid). The BP suggested a potentiometric titration method for determination of CTZ in its pure form; while it recommended an HPLC method for both cetirizine oral solution and tablets . Different analytical procedures were reported for its determination including HPLC [3,4,5,6], HPTLC , capillary electrophoresis  and spectrophotometry .
Phenylpropanolamine hydrochloride (PPA, Fig. 1b) is a nasal decongestant mainly used in combinations for relief of cold symptoms as it has indirect sympathomimetic activity . Its chemical name is (1RS, 2SR)-2-amino-1-phenylpropan-1-ol. The BP described non aqueous potentiometric titration for PPA . The USP suggested non-aqueous titration method using glacial acetic acid for PPA pure form and HPLC method for its capsules, extended released capsules, tablets, extended released tablets and oral solutions . There are different methods used for PPA determination as HPLC [5, 6, 11], capillary gas chromatography , spectrophotometry  and flow injection  methods.
Nimesulide (NMS, Fig. 1c) is a non-steroidal anti-inflammatory that acts by inhibition of COX-2 enzyme . It is 4′-nitro-2′-phenoxymethanesulphonanilide. The BP mentioned potentiometric titration method for NMS . The literature revealed several methods for NMS determination as HPLC [15,16,17], spectrophotometry  and TLC  methods.
The pharmaceutical preparation that contains the three drugs in a tablet dosage form is consisting of (5 mg CTZ, 25 mg PPA and 100 mg NMS) . The current study deals with two simple and sensitive methods for the simultaneous estimation of the three analytes included in this tablet preparation. The spectrophotometric method is a simple and sensitive cost-effective method. It doesn’t need any reagents or other tedious procedures. Although the literature contains two methods for the simultaneous determination of both CTZ and PPA [5, 6]; our proposed HPLC method is superior to the both mentioned methods. Despite Sunil et al.  provides an HPLC method for application in plasma and urine, it is less sensitive than our proposed method. Suryan et al. method  seeks from the disadvantage of longer retention times, and broader peaks. Our proposed HPLC method, consequently is more sensitive, rapid with sharper peaks than the other mentioned methods owing to the use of monolithic column through this study.
A Shimadzu (Kyoto, Japan) UV-1601 PC, UV–visible double-beam spectrophotometer was used. The third derivative spectra of the drugs were derived in the wavelength range (200–400) nm using Δλ = 8 nm and scaling factor = 10.
A Shimadzu LC-20 AD prominence liquid chromatograph (Japan) was used for HPLC analysis; with a Rheodyne injector valve and a SPD-20A UV detector set at wave length 215 nm.
Materials and reagents
Cetirizine hydrochloride pure sample was obtained from Apex Co., Cairo, Egypt (Batch No # 3003CZ8RJ) with 99.95% purity. Phenylpropanolmine hydrochloride (99.88% purity) was kindly brought from Cid Co., Egypt with Batch No # 41204. Nimesulide base was used with purity 99.90% as mentioned by the manufacturer, Batch No # 0006044. It is provided from Pharaonia Co., Alex, Egypt.
Organic solvents (HPLC grade) were purchased from Sigma-Aldrich (Germany).
Sodium hydroxide and sodium dihydrogen phosphate were purchased from ADWIC Co. (Egypt). Orthophosphoric acid (85%, w/v) was provided from Riedel-deHäen (Germany).
Chromolith® performance (RP-18 monolithic, 100 mm × 4.6 mm i.d.) is the column used for the investigation. The mobile phase used is a mixture of methanol and buffer (0.1 M phosphate buffer) in a ratio of (60:40 v/v) respectively. The pH was adjusted to be 7. The flow rate was 1 mL/min and the wavelength was 215 nm.
CTZ, PPA and NMS 400 µg/mL stock solutions were prepared by dissolving 40 mg of each the studied drugs in 100 mL methanol and further dilution was carried out to achieve the required concentrations for each of the two methods.
Construction of calibration graph
Serial dilutions of stock solutions were prepared to give concentrations of 8–90, 20–100 and 20–200 µg/mL for CET, PPA and NMS respectively. The third order derivative amplitudes were measured at 238, 218 and 305 nm for CTZ, PPA and NMS. A plot of the third derivative amplitude against the concentration was constructed to give the calibration curves.
CTZ, PPA and NMS working standard solutions were prepared by serial dilution of the stock solution in a 10 mL flask to obtain final concentration ranges; 5–40 µg/mL for CTZ, 10–100 µg/mL for PPA, and 10–120 µg/mL for NMS. The solutions were completed to the required volume by the mobile phase and were subjected to the chromatographic analysis under optimum conditions. Calibration graphs were constructed by plotting area under the curve against drug concentration in μg/mL [6,7,8].
Analysis of CTZ, PPA and NMS laboratory-prepared mixtures
Mixtures of CTZ, PPA and NMS in the ratio of 1:5:20 were prepared within the concentration ranges and analysed by the spectrophotometric strategy or the chromatographic strategy under the optimum conditions described in “Chromatographic conditions”. The percent recoveries were determined using regression equations or calibration graphs.
Analysis of CTZ, PPA and NMS in their co-formulated tablet
Laboratory co-formulated tablets were prepared as follows; accurately weighed 5 mg CTZ, 25 mg PPA and 100 mg NMS are mixed with 15 mg lactose, 10 mg magnesium stearate, 15 mg maize starch and 20 mg talc. One tablet was weighed, transferred to 100 mL volumetric flask, and completed to the mark with methanol. The solution undergoes 30 min sonication and then filtration till clear solution was obtained clear solution. Aliquots were taken within the concentration ranges for each drug (Table 1), and the chromatographic or spectrophotometric procedure was followed for calculating the percent recoveries .
Third derivative spectrophotometric method
The simultaneous analysis of the three drugs by classical spectrophotometric method is a challenge owing to the strong overlapping of their zero order spectra (Fig. 2), and the difference between their concentrations in the tablet. Also there was strong overlapping in first and second order derivative spectra, third derivative spectrophotometry was used in the analysis of the three drugs mixture without interference from each other (Fig. 3). CTZ could be assayed by measuring its third derivative amplitude at zero crossing points of NMS and PPA at 238 nm (Fig. 4) and PPA could be determined at zero crossing points of CTZ and NMS at 218 nm (Fig. 5). Also NMS was determined at zero crossing points of CTZ and PPA at 305 nm (Fig. 6).
Chromatographic method (HPLC)
Optimization of the chromatographic performance
Studying of chromatographic conditions was carried out to reach the optimum conditions that achieve good and efficient separation. Figure 7 shows typical chromatogram for CTZ, PPA and NMS laboratory-prepared mixture and Fig. 8 shows the typical chromatogram for laboratory prepared tablet.
Reversed-phase Chromolith® performance (RP-18 monolithic, 100 mm × 4.6 mm i.d.) and Promosil ODS 100 A column (250 × 4.6 mm i.d. 5 µm particle size) were tried during the separation. The first column was the suitable one as it resulted in well resolved peaks in shorter time.
Appropriate wavelength choice
The UV absorption spectra of the studied drugs in methanol show maxima at 211 and 231 nm for CTZ, 218 nm for PPA and 238, 296 and 307 nm for NMS (Fig. 2). HPLC chromatograms for studied drugs were scanned from 200 to 400 nm to determine the suitable wavelength and it was found that 215 nm was the suitable wavelength as the studied drugs showed high absorbance at this wavelength especially CTZ as it found in low concentration in the tablet dosage form.
Mobile phase composition
Different modifications were done for the mobile phase to enhance the efficiency of the separation procedures as illustrated in Table 2.
Type of organic modifier
Upon studying different organic solvents; it was found that acetonitrile and n-propanol showed overlapping between solvent peak and PPA giving split peak. Methanol was selected for optimum chromatographic conditions, as it gave higher number of theoretical plates with well resolved sharp peaks.
Ratio of organic modifier
The mobile phase which gives rapid separation of CTZ, PPA and NMS in good resolution is methanol: 0.1 M phosphate buffer in the ratio (60: 40, v/v). As the ratio of methanol increased the retention time of CTZ, PPA and NMS was decreased. The ratios 70 and 80% v/v of methanol caused overlapping between CTZ and NMS. CTZ band broadening was observed with ratio 50% (Table 2).
Ionic strength of phosphate buffer
0.1 M phosphate buffer was used as it gaves the highest number of theoretical plates with good resolution. Decreasing or increasing the ionic strength of phosphate buffer results in lower resolution or overlapping peaks.
Validation of the method
A linear relationship was established by plotting either the peak area or the derivative amplitude against the drug concentration in µg/mL for the HPLC and the spectrophotometric method respectively. The ranges of linearity were shown in Table 1. Equations referred to linear regression analysis are explained here:
Third derivative spectrophotometric method:
where: (3Dwavelength) is the third derivative amplitude of the spectra at the cited wavelength, and (C) is the concentration in µg/mL.
where: P is the peak area, C is the concentration of the drug in µg/mL and r is the correlation coefficient.
Theoretical basis assumes that the standard curve may be close to the origin, but practically it is rather difficult due to the presence of a reading for the solvent or the blank reading. As the intercept decreases in the calculations, this reflects that the solvent reading is almost near to zero . Linearity of the calibration curves was proved through statistical analysis  of the data (Table 1).
The limit of quantitation and limit of detection were calculated according to ICH recommendations .
where Sa is the standard deviation of the intercept of the calibration curve and b is the slope of the calibration curve. LOQ and LOD values for CTZ, PPA and NMS by the proposed methods were mentioned in Table 1.
In terms of accuracy; the results generated from the proposed methods were compared with those of well-established previous reports methods. The comparison method for CTZ and PPA describes reversed phase HPLC method  for simultaneous determination of both drugs using C18 column with UV detection at 217 nm. Concerning comparison method for determination of NMS; HPLC method  was utilized acetonitrile: 0.05M KH2PO4. The detection was carried out at 230 nm on C18 column. Accuracy was assessed through comparing the results of the proposed and the comparison methods and there was non-significant difference between the performance of them (Table 3). The ratio of CTZ, PPA and NMS in the tablet is not covered in the comparison method.
Repeatability and intermediate precision were tested to verify the precision of the proposed methods and the results were summarized in Table 4.
Robustness (for the HPLC method)
Some variables were changed on constancy of others for robustness investigation. These variables included; pH (6.9 ± 0.1) and phosphate buffer concentration (0.1 ± 0.005 M). These small changes had no effect on the separation and resolution of CTZ, PPA and NMS. This gave a good indication for the reliability of the proposed method.
Application in pharmaceutical preparations
Analysis of laboratory prepared mixtures
A successful determination for the three drugs in their laboratory prepared mixtures was performed and summarized in Table 5.
Dosage form analysis
Co-formulated tablets was also analyzed using the proposed HPLC and spectrophotometric methods as illustrated in Table 6. The results of statistical analysis were satisfactory as indicated by Student’s t test and variance ratio F test .
Third derivative spectrophotometry was used to analyze CTZ, PPA and NMS without interference from each other (Fig. 3). This method is simple, sensitive and efficient alternative to spectrophotometric methods mentioned for each of the three drugs in the literature [9, 13, 18], as it doesn’t need any reagents or additional time consuming steps.
The proposed approach also describes a novel HPLC method for the simultaneous determination of CTZ, PPA and NMS on a monolithic column. The established method is capable to separate the drugs with high efficacy and high resolution factor and within a short analysis time.
The current work provides the first method for the simultaneous analysis of CTZ, PPA and NMS in their pharmaceutical formulations. The developed spectrophotometric method is simple, rapid and economic. The HPLC method is a sensitive, reliable and time-saving method where separation of the studied analytes is achieved in less than 8 min. Moreover, the proposed methods overcome the analytical problems raised by the ratio of CTZ, PPA relative to NMS (1:5:20) and therefore could be used in the analysis of their co-formulated tablets in quality control laboratories.
international conference on harmonization
limit of quantification
limit of detection
Sweetman SC (ed) (2011) Martindale, the complete drug reference, 37th edn. The Pharmaceutical Press, London
British Pharmacopoeia (2013) Vol II, The stationary office, electronic version, London, pp 305, 1227, 1099
Karakuş S, Küçükgüzel I, Küçükgüzel SG (2008) Development and validation of a rapid RP-HPLC method for the determination of cetirizine or fexofenadine with pseudoephedrine in binary pharmaceutical dosage forms. J Pharm Biomed Anal 46:295–302
Moncrieff J (1992) Determination of cetirizine in serum using reversed-phase high-performance liquid chromatography with ultraviolet spectrophotometric detection. J Chromatogr B 583:128–130
Sunil K, Yogesh K, Abhijeet K (2013) Simultaneous estimation of cetirizine hydrochloride, phenylpropanolamine hydrochloride and paracetamol by RP-HPLC method. Int J Pharm Life Sci (IJPLS) 4:3122–3132
Suryan AL, Bhusari VK, Rasal KS, Dhaneshwar SR (2011) Simultaneous quantitation and validation of paracetamol, phenylpropanolamine hydrochloride and cetirizine hydrochloride by RP-HPLC in bulk drug and formulation. Int J Pharm Sci Drug Res 3:303–308
Makhija SN, Vavia PR (2001) Stability indicating HPTLC method for the simultaneous determination of pseudoephedrine and cetirizine in pharmaceutical formulations. J Pharm Biomed Anal 25:663–667
Azhagvuel S, Sekar R (2007) Method development and validation for the simultaneous determination of cetirizine dihydrochloride, paracetamol, and phenylpropanolamine hydrochloride in tablets by capillary zone electrophoresis. J Pharm Biomed Anal 43:873–878
Wankhede SB, Lad KA, Chitlange SS (2012) Development and validation of UV-spectrophotometric methods for simultaneous estimation of cetirizine hydrochloride and phenylephrine hydrochloride in tablets. Int J Pharm Sci Drug Res 4:222–226
The United States pharmacopeia 32, the National formulary 27, electronic version, US Pharmacopeial Convention, Rockville, 2009
Caraballoa I, Fernández-Arévaloa M, Holgadoa M, Alvarez-Fuentesa J, Rabascoa AM (1995) Communications simultaneous HPLC determination of some drugs commonly used in cold medications: dextromethorphan, dephenhydramine, phenylephrine, phenylpropanolamine and pseudoephedrine. Drug Dev Ind Pharm 21:605–613
Abbasia K, Bhangera MI, Khuhawar MY (2006) Capillary gas chromatographic determination of phenylpropanolamine in pharmaceutical preparation. J Pharm Biomed Anal 41:998–1001
Walash MI, El-Enany N, Saad S (2010) A new spectrophotometric method for determination of phenylpropanolamine HCl in its pharmaceutical formulations via reaction with 2, 3, 5, 6-tetrachloro-1,4-benzoquinone. Int J Biomed Sci 6:150–157
Viñas P, López-Erroz C, Cerdán FJ, Hernández-Córdoba M (1998) Determination of phenylpropanolamine and methoxamine using flow-injection with fluorimetric detection. Talanta 47:455–462
Nagaralli BS, Seetharamappa J, Gowda BG, Melwanki MB (2003) High-performance liquid chromatographic method for the determination of nimesulide in pharmaceutical preparations. J Anal Chem 58:778–780
Castoldi D, Monzani V, Tofanetti O (1988) Simultaneous determination of nimesulide and hydroxynimesulide in human plasma and urine by high-performance liquid chromatography. J Chromatogr B 425:413–418
Castoldi D, Monzani V, Tofanetti O (2003) Simultaneous determination of nimesulide and hydroxynimesulide in human plasma and urine by high-performance liquid chromatography. J Chromatogr B 58:778–780
Altinoza S, Dursunb OO (2000) Determination of nimesulide in pharmaceutical dosage forms by second order derivative UV spectrophotometry. J Pharm Biomed Anal 22:175–182
Pandya KK, Satia MC, Modi IA, Modi RI, Chakravarthy BK, Gandhi TP (1997) High-performance thin-layer chromatography for the determination of nimesulide in human plasma, and its use in pharmacokinetic studies. J Pharm Pharmacol 49:773–776
Miller JN, Miller JC (2005) Statistics and chemometrics for analytical chemistry, vol 107–149, 5th edn. Pearson Education Limited, Harlow, pp 39–73, 256
ICH harmonised tripartite guidelines. Validation of analytical procedures: text and methodology Q2(R1). http://www.ich.org/products/guidelines/quality/article/qualityguidelines.html. Accessed 01 Nov 2016
FA and NE planned and supervised the whole work. HM participated in the assay proposal, analysis and literature review. FA, NE and HM supervised the experimental work and participated in the assay design. AN carried out the practical part, collect the results and wrote the paper. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Aly, F.A., EL-Enany, N., Elmansi, H. et al. Simultaneous determination of cetirizine, phenyl propanolamine and nimesulide using third derivative spectrophotometry and high performance liquid chromatography in pharmaceutical preparations. Chemistry Central Journal 11, 99 (2017) doi:10.1186/s13065-017-0326-9
- Third derivative spectrophotometry
- Cetirizine (CTZ)
- Phenylpropanolamine (PPA)
- Nimesulide (NMS)