- Research article
- Open Access
Derivative spectrophotometric analysis of benzophenone (as an impurity) in phenytoin
Chemistry Central Journal volume 5, Article number: 85 (2011)
Three simple and rapid spectrophotometric methods were developed for detection and trace determination of benzophenone (the main impurity) in phenytoin bulk powder and pharmaceutical formulations. The first method, zero-crossing first derivative spectrophotometry, depends on measuring the first derivative trough values at 257.6 nm for benzophenone. The second method, zero-crossing third derivative spectrophotometry, depends on measuring the third derivative peak values at 263.2 nm. The third method, ratio first derivative spectrophotometry, depends on measuring the peak amplitudes of the first derivative of the ratio spectra (the spectra of benzophenone divided by the spectrum of 5.0 μg/mL phenytoin solution) at 272 nm. The calibration graphs were linear over the range of 1-10 μg/mL. The detection limits of the first and the third derivative methods were found to be 0.04 μg/mL and 0.11 μg/mL and the quantitation limits were 0.13 μg/mL and 0.34 μg/mL, respectively, while for the ratio derivative method, the detection limit was 0.06 μg/mL and the quantitation limit was 0.18 μg/mL. The proposed methods were applied successfully to the assay of the studied drug in phenytoin bulk powder and certain pharmaceutical preparations. The results were statistically compared to those obtained using a polarographic method and were found to be in good agreement.
Phenytoin is a hydantoin antiepileptic used to control partial and generalized tonic-clonic seizures. It has also been used in treatment of trigeminal neuralgia and cardiac arrhythmias . Benzophenone can be used as fixative for perfumes  and soaps, it prevents the ultraviolet (UV) light from damaging scents and colors in products, It can also be added to the plastic packaging as a UV blocker . Benzophenone used in the manufacture of antihistamines, hypnotics and insecticides  and so it considered as a potential impurity in diphenhydramine hydrochloride [4, 5], phenytoin bulk powder [4, 5] and phenytoin dosage forms [4, 5].
Benzophenone is irritating to the eyes, the skin and the respiratory tract [6, 7] and at excessive exposure, CNS disturbance and coma may be noted . Chhabra  and Rhodes et al.  found that, the liver is a primary target organ of benzophenone toxicity in rats and mice. The toxicological profile of benzophenone is similar to a number of known hepatocarcinogens, suggesting that benzophenone is a potential liver carcinogen. The kidney was also identified as a target organ of benzophenone toxicity. Estrogenic and antiandrogenic activities of benzophenone were comparatively examined with hormone-responsive reporter assay in various cell lines [10–12].
The United States Pharmacopoeia USP  recommended HPLC procedures for the determination of benzophenone as a main impurity in phenytoin. The BP  specifies a TLC method to limit benzophenone in phenytoin.
A review of the literature revealed that only one (polarographic) method has been published for the determination of benzophenone, as an impurity, in phenytoin . Although those methods offer a high degree of specificity, the instrumentation limitations preclude their use in routine analysis. To the best of our knowledge, no spectrophotometric methods have been yet described for the determination of benzophenone, as an impurity, in phenytoin. Therefore, it was desirable to develop a simple and fast procedure that could be applied in quality control laboratories for the determination of benzophenone, as an impurity, in phenytoin.
Derivative spectrophotometry offers greater selectivity than does normal spectrophotometry in the simultaneous determination of two or more compounds without previous chemical separation [14–16]. The principles and advantages of this technique have been described by O'Haver and Green .
Ratio derivative spectrophotometry is based on the use of the first derivative of the ratio spectra. This method was developed by Salinas et al . Berzas Nevado et al. extended this method to resolving ternary mixtures .
The present paper describes simple and rapid methods for the determination of benzophenone, as impurity, in phenytoin bulk powder and certain pharmaceutical preparations by zero-crossing first derivative spectrophotometry, zero-crossing third derivative spectrophotometry and ratio first derivative spectrophotometry without prior separation.
Spectrophotometric analysis was carried out on a Shimadzu (Kyoto, Japan) UV-1601 PC, UV-Visible double-beam spectrophotometer with matched 1-cm pathlength quartz cells. Suitable settings were: Slit width, 1 nm; scan speed, fast; sampling interval, auto.
The first derivative spectra of benzophenone were derived in the wavelength range (250-350) nm using Δλ = 2 nm and scaling factor = 100.
The third derivative spectra of the studied drug were derived in the wavelength range (240-300) nm using Δλ = 8 nm and scaling factor = 2000.
The ratio derivative spectra were derived in the wavelength range (200-298) nm using Δλ = 8 nm and scaling factor = 1 for smoothing of ratio spectra and Δλ = 4 nm for the first derivative of ratio spectra with scaling factor = 10.
Materials and reagents
All the chemicals used were of Analytical Reagent grade and the solvents were of spectroscopic grade.
Benzophenone was kindly provided by (El-Nasr Pharmaceutical Chemicals Company (ADWIC), Egypt). The purity of the sample was found to be 100.15 ± 0.36 which was determined by applying the reported polarographic(13) method.
Phenytoin (Aldrich Chemical Co., Ltd., Dorset, England).
Phenytin® ampoules (labeled to contain 100 mg phenytoin sodium/2 mL ampoule, batch # 95040) were manufactured in The Nile Co. for Pharmaceuticals & Chemical Industries, Cairo, Egypt.
Phenytin® capsules (labeled to contain 100 mg phenytoin sodium/capsule, batch # 36050) were manufactured in The Nile Co. for Pharmaceuticals & Chemical Industries, Cairo, Egypt.
Ipanten® capsules (labeled to contain 100 mg phenytoin sodium/capsule, batch # AC1009) were manufactured in ACDIMA International trading, Egypt.
All Pharmaceutical preparations were purchased from commercial sources.
Methanol (Prolabo, France).
Stock solutions (1.0 mg/mL) of each of benzophenone and phenytoin were prepared by dissolving 10.0 mg of each in methanol and completed to 10 mL with the same solvent. Working standard solutions (100.0 μg/mL) were prepared by subsequent dilution to 25 mL with methanol. An additional working solution (10.0 mg/mL) of phenytoin was prepared by the same manner, to be used in the preparation of synthetic mixtures. The solutions were stable for at least 7 days when kept in the refrigerator at 4°c.
Construction of the Calibration Graphs
Daily working standard solutions were prepared from the previously prepared working standard solution by serial dilutions with methanol to contain 1 - 10 μg/mL for benzophenone and to contain 200 μg/mL for phenytoin (final concentrations). The zero order absorption spectra were recorded against a reagent blank (methanol).
The absolute values of the first order derivatives were obtained by zero-crossing technique with measurements at 257.6 nm for benzophenone.
For the determination by third derivative spectrophotometry, the absolute values of the third derivative were measured at 263.2 nm.
For the determination by ratio derivative spectrophotometry, the first derivative of the ratio spectra (the spectra of benzophenone divided by the spectrum of 5.0 μg/mL phenytoin solution) was recorded. The amplitudes at 272 nm were measured.
The derivative amplitudes were then plotted against the final concentrations to get the calibration graphs. Alternatively, the corresponding regression equations were derived.
Application of the Proposed Methods to the Analysis of Benzophenone in Prepared Synthetic Mixtures
Accurately measured aliquots of the suitable working standard solutions of both drugs were transferred into a series of 10-mL volumetric flasks to prepare different synthetic mixtures of benzophenone and phenytoin in the ratios of 0.5% (the BP  limits of benzophenone in phenytoin) and 0.1% (the USP  limits of benzophenone in phenytoin). The solutions were then diluted with methanol to volume, mixed and aliquots containing suitable concentrations were taken and analyzed as described under construction of the calibration graphs. The concentration of benzophenone was determined using, either the calibration curve or the corresponding regression equation.
Application of the Proposed Methods to the Analysis of Added Benzophenone in Phenytoin Dosage Forms
An accurately weighed quantity of the mixed contents of 10 capsules equivalent to 0.5 g of phenytoin was transferred into a small conical flask, and then extracted by sonication for 15 minutes with 15 mL of methanol and filtering into 50-mL volumetric flask. Repeat the extraction with further two 15-mL portions of methanol. The conical flask was washed with several milliliters of methanol. The washings were passed into the same volumetric flask and completed to the volume with the same solvent. The above procedure under application of the proposed methods to the analysis of benzophenone in prepared synthetic mixtures was then followed. The nominal content was calculated either from the previously plotted calibration graph or using the corresponding regression equation.
An accurately measured volume of mixed content of 10 ampoules equivalent to 1.0 g of phenytoin base transferred into 100-mL volumetric flask, and then diluted to the mark with methanol. The above procedure under application of the proposed methods to the analysis of benzophenone in prepared synthetic mixtures was then followed. The nominal content was calculated either from the previously plotted calibration graph or using the corresponding regression equation.
Results and discussion
The BP  limits are 0.5% for benzophenone in bulk powder and in, capsules and ampoules of phenytoin, whereas, the USP  limits are 0.1% for the studied drug in phenytoin bulk powder. Moreover the specific absorbance of phenytoin (the major component) is higher than the specific absorbance of benzophenone (the minor component). So, the analysis of such mixture by conventional spectrophotometry was challenging. Therefore, we resorted to first derivative, third derivative and ratio first derivative spectrophotometry in an attempt to detect and determine benzophenone (the main impurity) in phenytoin.
Dilution with different solvents such as 0.1 M HCl, 0.1 M NaOH, methanol and acetonitrile were tried for performance investigations. It was found that; methanol was the best solvent for dilution as it gave the highest sensitivity with good shaped spectrum.
Figure 1(a, b) shows the absorption spectra of benzophenone and phenytoin in methanol which overlap sufficiently to demonstrate the resolving power of the proposed methods.
'Zero-crossing' first derivative spectrophotometry
Because the derivative spectrophotometric technique enhances the detectability of the minor features of the UV absorption spectrum, the first derivative spectra of both benzophenone and phenytoin (Figure 2) displays features which may permit more specific and selective determination of the benzophenone in the presence of phenytoin. The zero-crossing method is the most common procedure for conducting analytical calibration in derivative spectrophotometry, so benzophenone was determined by measurement of its first derivative amplitude at the zero-crossing of phenytoin at 257.6 nm (1D257.6; order derivativeDerivativewavelength measure).
Figure 2 shows the first derivative spectra of benzophenone in concentration range of 1-10 μg/mL with constant phenytoin concentration (200 μg/mL).
'Zero-crossing' third derivative spectrophotometry
Second and fourth derivative spectra were scanned but a reproducible zero crossing point could not be found. Figure 3 shows the third derivative spectra of benzophenone which could be determined by measurement of its third derivative amplitude at the zero-crossing of phenytoin at 259 nm (3D259), or at 263.2 nm (3D263.2) both with the same sensitivity.
Ratio first derivative spectrophotometry
Figure 4 shows the ratio spectra of different concentrations of benzophenone (spectra divided by the spectrum of a 5 μg/mL of phenytoin) while, Figure 5 shows their first derivatives. As it can be seen, the amplitude at 272 nm (1DD272; order derivativeDerivative Dividedwavelength measure) in the ratio derivative spectra corresponds to benzophenone present in the solution, so it can be used for its quantitative determination. The influence of Δλ for obtaining the first derivative of the ratio spectra was tested to obtain the optimum wavelength interval; Δλ = 4 nm was considered as suitable for this. For selecting the standard solution as divisor, different concentrations were tested and different calibration curves were obtained.
The best results in terms of signal-to-noise ratio, sensitivity and repeatability were obtained by using the spectrum of 5 μg/mL phenytoin solution as divisor in the determination of benzophenone.
To ensure the validity of the proposed methods, the results were compared with those of a comparison differential pulse (DPP) polarographic method . However USP  recommends an HPLC method as limit test for benzophenone in bulk drug only, while BP  recommends a thin-layer (TLC) chromatographic method as a limit test for benzophenone in phenytoin bulk powder and dosage forms.
Linearity and Range
The calibration graphs for the determination of benzophenone by the proposed methods were constructed by plotting the derivative amplitudes versus the concentrations as shown in Figures 6, 7 and 8. The graphs were found to be rectilinear over the concentration ranges cited in Table 1.
Statistical analysis  of the data gave high values of correlation coefficients (r) of the regression equations, small values of the standard deviations of residuals (S y/x ), of intercept (S a ), and of slope (S b ), and small values of percentage relative standard deviation and percentage relative error (Table 1). These data proved the linearity of the calibration graphs and the agreement of the results with Beer's law.
Accuracy and Precision
To prove the accuracy of the proposed methods, the results of the assay of benzophenone as impurities in phenytoin bulk powder and its pharmaceutical preparations were compared with those of a reported comparison polarographic method . Moreover, several synthetic mixtures in BP  and USP  limits ratio were also analyzed. Statistical analysis  of the results obtained by the proposed and comparison method using Student's t-test (t) and variance ratio F-test (F) showed no significant differences between them regarding accuracy and precision, respectively (Tables 2, 3 and 4). Intraday (repeatability) and inter-day (intermediate) precisions were assessed using three concentrations and three replicates of each concentration. The relative standard deviations were found to be very small indicating reasonable repeatability and intermediate precision of the proposed methods (Table 5).
The specificity of each of the proposed methods was investigated by observing any interference encountered from common dosage form excipients. It was shown that these compounds did not interfere with the results of the proposed methods (Table 4).
Limit of Detection (LOD) and Limit of Quantitation (LOQ)
The limit of quantitation (LOQ) was determined by establishing the lowest concentration that can be measured according to ICH Q2)R1) recommendation  below which the calibration graph is non linear while the limit of detection (LOD) was determined by evaluating the lowest concentration of the analyte that can be readily detected. The results are shown in Table 1. LOQ and LOD were calculated according to the following equations :
Where Sa is the standard deviation of the intercept of regression line, and b is the slope of the calibration curve.
To examine the ruggedness of the procedures, the intraday and inter-day precisions were evaluated as shown in Table 5. The precisions of the proposed methods were fairly high, as indicated by the low values of percentage relative standard deviation (% RSD) for the studied drugs.
Analysis of Synthetic Mixtures
The proposed methods were successfully applied to the analysis of benzophenone in synthetic mixtures containing the BP  and USP  limits of benzophenone in phenytoin. The average percent recoveries were based on the average of three replicate determinations. The results obtained were in good agreement with those obtained by the comparison polarographic method . The results are shown in Table 3.
Analysis of Dosage Forms
The proposed methods were successfully applied to the assay of benzophenone (by spiking) in phenytoin ampoules and capsules. The average percent recoveries of different concentrations were based on the average of three replicate determinations. The results obtained showed non-significant differences with those obtained by the comparison polarographic method  as shown in Table 4.
The proposed methods could be utilized for the detection and determination of benzophenone, as impurity, in phenytoin bulk powder and its pharmaceutical preparations using simple, reliable, rapid, and economical procedures. The zero-crossing derivative spectrophotometry is more rapid and simple than ratio derivative spectrophotometry; however the ratio derivative spectrophotometry has greater sensitivity and accuracy. The use of standardized spectra as divisors minimizes experimental errors and background noise.
These proposed methods could be regarded as useful alternative to the official USP chromatographic, the BP qualitative TLC and reported polarographic methods in the routine quality control of pharmaceutical formulations, allowing qualitative and quantitative determinations to be simultaneously and rapidly achieved with a relatively inexpensive instrumentation.
Sweetman S: Martindale: The Complete Drug Reference. 2007, The Pharmaceutical Press: London, 447-453. 35
The Merck Index on CD-Rom. Edited by: Budavari S. 1996, By Merck & Co., Inc., White House Station, NJ, USA, 1129. 12
Benzophenone-Wikipedia, the free encyclopedia. (on line) at [http://en.wikipedia.org/wiki/Benzophenone] The page was last modified on 20 April 2009 (accessed at 04 May 2009)
British Pharmacopoeia Commission: The British Pharmacopoeia 2007. 2007, The Stationery Office: London, Electronic version
United States Pharmacopeial Convention: The United States Pharmacopoeia 30, the National Formulary 25. 2007, Rockville, MD, USA: United States Pharmacopeial, Electronic version
International Chemical Safety Cards for benzophenone. (on line) at [http://www.cdc.gov/niosh/ipcsneng/neng0389.html]
Sigma-Aldrich Catalogue. (on line) at [www.sigmaaldrich.com]
Chhabra RS: National Toxicology Program (NTP) Toxicity Studies of Benzophenone (CAS No. 119-61-9). Administered in Feed to rats and mice. Toxicol Rep Series. 2000, 61: 6-
Rhodes MC, Bucher JR, Peckham JC, Kissling GE, Hejtmancik MR, Chhabra RS: Carcinogenesis studies of benzophenone in rats and mice. Food and Chemical Toxicology. 2007, 45: 843-851. 10.1016/j.fct.2006.11.003.
Suzuki T, Kitamura S, Khota R, Sugihara K, Fujimoto N, Ohta S: Estrogenic and antiandrogenic activities of 17 benzophenone derivatives used as UV stabilizers and sunscreens. Toxicol Appl Pharmacol. 2005, 203: 9-17. 10.1016/j.taap.2004.07.005.
Hayashi T, Okamoto Y, Ueda K, Kojima N: Formation of estrogenic products from benzophenone after exposure to sunlight. Toxicol Lett. 2006, 167: 1-7. 10.1016/j.toxlet.2006.08.001.
Kato T, Tada-Oikawa S, Takahashi K, Saito K, Wang L, Nishio A, Hakamada-Taguchi R, Kawanishi S, Kuribayashi K: Endocrine disruptors that deplete glutathione levels in APC promote Th2 polarization in mice leading to the exacerbation of airway inflammation. Eur J Immunol. 2006, 36: 1199-1209. 10.1002/eji.200535140.
Razak OA, Gazy AA, Wahbi AM: Polarographic determination of phenytoin and benzophenone (as impurity) in pharmaceutical preparations. J Pharm Biomed Anal. 2002, 28: 613-619. 10.1016/S0731-7085(01)00669-0.
Morelli B: Zero-crossing derivative spectrophotometry for the determination of mixtures of cephaloridine and cephalothin in pure and dosage forms. J Pharm Sci. 1988, 77: 615-621. 10.1002/jps.2600770713.
Morelli B: Second-derivative spectrophotometric assay of mixtures of dicloxacillin sodium and ampicillin sodium in pharmaceuticals. J Pharm Sci. 1988, 77: 1042-1046. 10.1002/jps.2600771211.
Morelli B: Determination of a quaternary mixture of vitamins B6, B1, and B12 and uridine 5'-triphosphate, by derivative spectrophotometry. J Pharm Sci. 1995, 84: 34-37. 10.1002/jps.2600840109.
O'Haver TC, Green GL: Numerical error analysis for derivative spectrometry for the quantitative analysis of mixtures. Anal Chem. 1976, 48: 312-318. 10.1021/ac60366a016.
Salinas F, Berzas Nevado JJ, Espinosa Mansilla A: A new spectrophotometric method for quantitative multicomponent analysis resolution of mixtures of salicylic and salicyluric acids. Talanta. 1990, 37: 347-351. 10.1016/0039-9140(90)80065-N.
Berzas Nevado JJ, Guiberteau Cabanillas C, Salinas F: Spectrophotometric resolution of ternary mixtures of salicylaldehyde, 3-hydroxybenzaldehyde and 4-hydroxybenzaldehyde by the derivative ratio spectrum-zero crossing method. Talanta. 1992, 39: 547-553. 10.1016/0039-9140(92)80179-H.
Miller JN, Miller JC: Statistics and Chemometrics for Analytical Chemistry. 2005, Pearson Education Limited: Harlow, England, 107-149. 256, 5
ICH Harmonized Tripartite Guideline Validation of Analytical Procedures: Text and Methodology, Q2(R1), Current Step 4 Version, Parent Guidelines on Methodology Dated November 6, 1996, Incorporated in November 2005. [http://www.ich.org/fileadmin/Public_Web_Site/ICH_Products/Guidelines/Quality/Q2_R1/Step4/Q2_R1__Guideline.pdf] (accessed February 28, 2008)
We gratefully acknowledge the financial support received from Faculty of pharmacy, Mansoura University, Egypt.
The authors declare that they have no competing interests.
MIW designed the proposed method and analyzed the data statistically. MSR proposed, planned and supervised the whole work. ZAS coordinated the study and modified the text. MMS carried out the experimental work. All authors read and approved the final manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.