- Research article
- Open Access
Development and validation of a repharsed phase- HPLC method for simultaneous determination of rosiglitazone and glimepiride in combined dosage forms and human plasma
© El-Enany et al 2011
- Received: 7 August 2011
- Accepted: 26 January 2012
- Published: 26 January 2012
Rosiglitazone (ROZ) and glimepiride (GLM) are antidiabetic agents used in the treatment of type 2 diabetes mellitus. A survey of the literature reveals that only one spectrophotometric method has been reported for the simultaneous determination of ROS and GLM in pharmaceutical preparations. However the reported method suffers from the low sensitivity, for this reason, our target was to develop a simple sensitive HPLC method for the simultaneous determination of ROZ and GLM in their combined dosage forms and plasma.
A simple reversed phase high performance liquid chromatographic (RP-HPLC) method was developed and validated for the simultaneous determination of Rosiglitazone (ROS) and Glimepiride (GLM) in combined dosage forms and human plasma. The separation was achieved using a 150 mm × 4.6 mm i.d., 5 μm particle size Symmetry® C18 column. Mobile phase containing a mixture of acetonitrile and 0.02 M phosphate buffer of pH 5 (60: 40, V/V) was pumped at a flow rate of 1 mL/min. UV detection was performed at 235 nm using nicardipine as an internal standard. The method was validated for accuracy, precision, specificity, linearity, and sensitivity. The developed and validated method was successfully used for quantitative analysis of Avandaryl™ tablets. The chromatographic analysis time was approximately 7 min per sample with complete resolution of ROS (tR = 3.7 min.), GLM (tR = 4.66 min.), and nicardipine (tR, 6.37 min). Validation studieswas performed according to ICH Guidelines revealed that the proposed method is specific, rapid, reliable and reproducible. The calibration plots were linear over the concentration ranges 0.10-25 μg/mL and 0.125-12.5 μg/mL with LOD of 0.04 μg/mL for both compounds and limits of quantification 0.13 and 0.11 μg/mL for ROS and GLM respectively.
The suggested method was successfully applied for the simultaneous analysis of the studied drugs in their co-formulated tablets and human plasma. The mean percentage recoveries in Avandaryl™ tablets were 100.88 ± 1.14 and 100.31 ± 1.93 for ROS and GLM respectively. Statistical comparison of the results with those of the reference method revealed good agreement and proved that there were no significant difference in the accuracy and precision between the two methods respectively. The interference likely to be introduced from some co-administered drugs such as glibenclamide, gliclazide, metformine, pioglitazone and nateglinide was investigated.
- rosiglitazone (ROS)
- glimepiride (GLM)
- human plasma
Regarding glimepiride (GLM), 3-ethyl-4-methyl-N-(4-[N-((1r,4r)-4 methylcyclohexylcarbamoyl)sulfamoyl]phenethyl)-2-oxo-2,5-dihydro-1H-pyrrole-1-carboxamide (Figure 1) it is a medium-to-long acting sulfonylurea anti-diabetic drug. It is the first of the third-generation sulfonylurea, and is very potent .
The two drugs are co-formulated commercial tablets which is used for the treatment of patients with type 2 diabetes:
Literature survey reveals few analytical methods for the determination of rosiglitazone (ROS) in pharmaceutical preparations and biological fluids, including: spectrophotometry [3, 4], voltammetry  and HPLC [6–8].
To the best of our knowledge in recent years only one spectrophotometric method has been reported for the simultaneous determination of ROS and GLM in pharmaceutical preparations . However the reported method suffers from the low sensitivity, for this reason, our target was to develop a simple sensitive HPLC method for the simultaneous determination of ROZ and GLM in their combined dosage forms and plasma. Because HPLC methods have been widely used for routine quality-control assessment of drugs, because of their sensitivity, repeatability, and specificity, we have developed a simple and specific RP-HPLC method for simultaneous determination of ROS and GLM in pharmaceutical dosage forms. Because analytical methods must be validated before use by the pharmaceutical industry, the proposed HPLC-UV method was validated in accordance with International Conference on Harmonization (ICH) guidelines , by assessing its selectivity, linearity, accuracy, precision, and limits of detection and quantitation.
Chromatographic System and Conditions
The HPLC system is LC-10A (Shimadzu, Kyoto, Japan) consisted of a pump (LC-10), UV detector (SPD-10AV) and auto-injector (SIL-10 AXL).
Column: Symmetry® C18 (150 mm × 4.6 mm i.d., 5 μm particle size), Waters, Ireland. The column hold up value was the first deviation of the base line obtained.
Mobile phase: consists of a mixture of acetonitrile and 0.02 M phosphate buffer of pH 5 (60: 40, V/V). The run time was 8 min at a flow rate was 1 ml min-1., detector wavelength 235 nm. The injection volume was 30 μl.
Materials and reagents
All materials and reagents used were of Analytical Reagent grade.
- Rosiglitazone maleate reference standard was kindly supplied by Glaxo SmithKline, Egypt
- Glimepiride reference standard was kindly supplied by Delta Pharma, Egypt.
- Avandaryl™ tablets; labeled to contain 4 mg rosiglitazone maleate and 2 mg glimepride per tablet, were purchased from commercial sources in the local market.
- Acetonitrile, methanol, ethanol and propanol (Sigma-Aldrich, Germany).
- Sodium dihydrogen phosphate and disodium hydrogen phosphate (Winlab, England).
- Human plasma was kindly provided by Mansoura University Hospitals, Mansoura, Egypt and was kept frozen until used after gentle thawing.
Stock solutions of ROS, GLM and nicardipine were prepared by dissolving 20 mg of each pure substance in 100 mL of methanol to obtain a stock solution of 200 μg/mL. These stock solutions were further diluted with the same solvent and then with the mobile phase as appropriate to obtain the working standard solution of 0.10-25 μg/mL and 0.125-12.5 μg/mL for ROS, GLM respectively). The solutions were stable for at least three days without alteration when kept in the refrigerator.
General procedures for analysis of laboratory prepared mixture of ROS and GLM
Aliquot volumes containing ROS, GLM and nicardipine (IS) were quantitatively transferred into a series of 10 mL volumetric flasks, so that the final concentration were in the range of 0.10-25, 0.125-12.5 and 20 μg/mL for ROS, GLM and NIC (IS) respectively. The solutions were completed to the volume with the mobile phase. 30 μL aliquots were injected (in triplicate) and the calibration curve was constructed by plotting the peak area ratio against the final concentration of the drug (μg/mL). Alternatively the corresponding regression equations were derived.
Application of the proposed method to the analysis of Avandaryl™ tablets
Ten Avandaryl™ tablets were weighed and pulverized well. A weighed quantity of the powder equivalent to 20.0 mg of ROS and 10 mg of GLM were transferred into a small conical flask and extracted three successive times each with 30 mL of methanol. The extracts were filtered into 100 mL volumetric flask. The conical flask and filter paper were washed with few milliliters of methanol. The washings were passed into the same volumetric flask, and completed to volume with same solvent. Aliquots covering the working concentration ranges were transferred into 10 ml volumetric flasks. The procedures described under "General procedures" were performed. The nominal contents of the tablets were calculated using the corresponding regression equations.
Application of the proposed method to spiked human plasma
Blank plasma was spiked with standard solutions of both drugs to achieve the following calibration standard concentrations 0.1- 0.25- 0.5- 0.75 and 1.0 μg/mL; for both of ROS and GLM respectively at constant concentration of NIC (2 μg mL-1).
Aliquots of human plasma (1 mL) spiked with increasing concentrations of the drugs were transferred into centrifugation tubes. Acetonitrile (2 ml) was added to each tube. After vortex mixing for 5 min., the mixtures were centrifuged at 3500 rpm for 15 min. at room temperature. The Supernatant was evaporated using nitrogen gas in a heating block set at 45°C. Dried samples were reconstituted with the mobile phase and. Aliquots of 30 μL were injected (triplicate) and eluted with the mobile phase under the reported chromatographic conditions. A blank experiment was carried out simultaneously. The peak area ratio was plotted versus the concentration of the drugs in μg mL-1 to get the calibration graph. Alternatively, the regression equations were derived.
Chromatographic condition and optimization
Choice of column
Trials were performed using
Symmetry® C18 column (150 mm × 4.6 mm i.d., 5 μm particle size), Waters, Ireland; Zorbax SB-Phenyl column (250 mm × 4.6 mm i.d., 5 μm particle size), Agilent Technologies, USA, and Hypurity C8 column (250 mm × 4.6 mm i.d., 5 μm particle size). Thermo Electron Corporation. The experimental studies revealed that the first column was the best one, since it produced symmetrical peaks with high resolution. The phenyl column resulted in separation of both drugs but with low resolution for ROS (tR = 3.1 min.) and GLM (tR = 3.69 min.). On the other hand using C8 column resulted in separation of both drugs but with low sensitivity for ROS (tR = 3.4 min.) and GLM (tR = 4.2 min.)
Mobile phase composition
Several modifications in the mobile phase composition were performed in order to study the possibilities of improving the performance of the chromatographic system. These modifications included: the type and ratio of the organic modifier, effect of flow rate and the pH
Type of organic modifier
Acetonitrile was replaced by methanol, ethanol and propanol. Although ethanol and propanol gave sharp peaks, yet they were not selected as using ethanol gave longer retention time and propanol resulted in poor resolution. Methanol on the other hand gave very small and broad peaks. In addition, the three organic modifiers showed low sensitivity. So, acetonitrile was the organic modifier of choice giving symmetrical highly sensitive peaks in a reasonable time.
Ratio of organic modifier
Effect of pH
As it is well known that the buffer is important during the liquid chromatographic phase because it allows for the separation of organic compounds, 0.02 M phosphate buffer was chosen a buffer as it is very handy with better solubility in organic solvents.
The effect of flow rate on the formation and separation of peaks of the studied compounds was studied and a flow rate of 1 mL/min. was the optimal for good separation in a reasonable time.
Validation of the method
Analytical performance data for the determination of the studied drugs by the proposed method.
Linearity range (μg/mL)
-5.9 × 10-4
-1.7 × 10-3
Correlation coefficient (r)
S.D. of residuals (S y/x )
6.90 × 10-3
2.68 × 10-3
S.D. of intercept (S a )
1.58 × 10-3
1.39 × 10-3
S.D. of slope (S b )
2.88 × 10-4
2.36 × 10-4
Percentage relative standard deviation, % RSD
Percentage relative error, % Error
Limit of detection, LOD (μg/mL)
Limit of quantitation, LOQ (μg/mL)
Where: PA is the peak area ratio, C is the concentration of the drug in μg/ml and r is the correlation coefficient.
Limit of quantitation (LOQ) and limit of detection (LOD)
Where Sa is the standard deviation of the intercept of the regression lines and b is the slope of the calibration curve.
LoQ values were found to be 0.13 and 0.11 μg/mL for ROS and GLM, respectively, while LoD values were found to 0.04 μg mL-1 for both compounds.
Statistical evaluation of the regression lines regarding standard deviation of the residual (Sy/x), standard deviation of the intercept (Sa) and standard deviation of the slope (Sb) is shown in Table 1. The small values of the figures point out to the low scattering of the points around the calibration curve (Table 1) thus indicating high precision of the method .
The proposed method was evaluated by studying the accuracy as percent relative error (%Er) and precision as percent relative standard deviation (%RSD). The results are abridged in Table 1.
The accuracy of analytical method is defined as the similarity of the results obtained by this method to the true value and the precision as the degree of that similarity .
Assay results for the determination of the studied drugs in their synthetic mixture and co-formulated tablets.
Concentration taken (μg/mL)
Concentration found (μg/mL)
Reference method 
1- Synthetic mixture(a)
2-AvandarylTM tablets(b) (4 mg ROS + 2 mg GLM/tablet)
The proposed procedure offers additional advantages over the reference procedure  in that the former is more sensitive with good accuracy and precision. The latter method depends on the analysis of ROS-GLM binary mixture by first derivative spectrophotometric technique up to the concentration of 30 μg/mL.
Dosage form analysis
The proposed method was further applied to the determination of the studied drugs in their co-formulated tablets. The results shown in Table 2 are in good agreement with those obtained with the reference method . Statistical analysis of the results obtained using Student's t-test and variance ratio F-test  revealed no significant difference between the performance of the two methods regarding the accuracy and precision, respectively (Table 2).
The selectivity of the method was investigated by observing any interference encountered from common tablet excipients such starch, magnesium stearate, lactose and avisil. It was shown that these compounds did not interfere with the results of the proposed method.
The high sensitivity of the proposed method allowed the determination of rosiglitazone in spiked human plasma. The absolute bioavailability of rosiglitazone is 99%. Following single oral administration of rosiglitazone 8 mg to fasting diabetic patient, a mean peak plasma concentration of 0.117 μg/mL was observed . This concentration lies within the linear concentration range of the proposed method.
Following oral administration of a single dose 2 mg of glimipride, a mean plasma level of 0.176 μg/mL was achieved at 2.1 hour .
The interference arising from the endogenous amino acids has been overcome by simple precipitation of protein. The method just involved simple plasma protein precipitation with acetonitrile prior to measurement .
Application of the proposed method to the determination of the studied drugs in spiked human plasma.
Concentration taken (μg/mL)
Concentration found (μg/mL)
for GLM respectively
Where: PA is the peak area ratio, C is the concentration of the drug in μg/mL and r is the correlation coefficient.
The high value of the correlation coefficient (r) indicates the good linearity of the calibration graph.
Precision of the proposed method in spiked human plasma
Validation data for the HPLC analysis of ROS and GLM in spiked human plasma.
Concentration added (μg/mL)
Concentration found (μg/mL)
Effect of co-administered drugs such as pioglitazone, nateglinide, metformine, glibenclamide and gliclazide was studied. All of the studied drugs interfered with the peak of either ROS or GLM. This necessities developing other separation method.
The proposed HPLC method is applicable for the simultaneous separation and determination of ROS and GLM in their co-formulated dosage forms. It has distinct advantages over other existing method regarding sensitivity, time saving and minimum detection limits. Moreover, it can be applied to the determination of both drugs in spiked human plasma.
- Balfour JA, Plosker GL: Rosiglitazone. Drugs. 1999, 57 (6): 921-30. 10.2165/00003495-199957060-00007.View ArticleGoogle Scholar
- AVANDARYL® (rosiglitazone maleate and glimepiride) Tablets. [http://www.medicineonline.com/drugs/A/4200/AVANDARYL-rosiglitazone-maleate-and-glimepiride-Tablets.html]
- Gomes P, Steppe M: First-derivative spectrophotometry in the analysis of rosiglitazone in coated tablets. J AOAC Int. 2006, 89: 1296-1299.Google Scholar
- Walash MI, El-Brashy A, El-Enany N, Kamel ME: Spectrofluorimetric and spectrophotometric determination of rosiglitazone maleate in pharmaceutical preparations and biological fluids. Pharmaceutical Chemistry Journal. 2009, 43: 697-709. 10.1007/s11094-010-0383-z.View ArticleGoogle Scholar
- El Sherbiny D, El-Enany N, Belal F: Voltammetric Determination of Rosiglitazone in Pharmaceutical Preparations and Human Plasma. Anal Lett. 2008, 41: 806-821. 10.1080/00032710801935046.View ArticleGoogle Scholar
- Ali AR, Duraidi II, Saket MM, Abu-Nameh ES: Column high-performance liquid chromatographic method for the simultaneous determination of rosiglitazone and metformin in a pharmaceutical preparation. J AOAC Int. 2009, 92: 119-124.Google Scholar
- Kang X, Wang F, Xie Z, Li H: A high performance liquid chromatography method for simultaneous determination of rosiglitazone and gemfibrozil in human plasma. J Chromatogr B Analyt Technol Biomed Life Sci. 2009, 877: 645-648. 10.1016/j.jchromb.2009.01.001.View ArticleGoogle Scholar
- Yu J, Ma K, Qi J, Jin G, Wang Y, Fang S, Li G: Bioequivalence and comparison of pharmacokinetic properties of 4-mg tablet formulations of rosiglitazone hydrochloride and rosiglitazone maleate: a single-dose, randomized, open-label, two-period crossover study in healthy adult male Chinese volunteers. Clin Ther. 2008, 30: 2272-2279. 10.1016/j.clinthera.2008.12.011.View ArticleGoogle Scholar
- Altinöz S, Tekeli D: Analysis of glimepiride by using derivative UV spectrophotometric method. J Pharm Biomed Anal. 2001, 24: 507-515. 10.1016/S0731-7085(00)00445-3.View ArticleGoogle Scholar
- Ma HL, Xu MT, Qu P, Ma XH: Polarographic behavior and determination of glimepiride. Yao Xue Xue Bao. 2005, 40: 750-753.Google Scholar
- A K, G S, C MR, Bhat K, A R, P M, M S, K K, N U: Simultaneous determination of pioglitazone and glimepiride in bulk drug and pharmaceutical dosage form by RP-HPLC method. Pak J Pharm Sci. 2008, 21: 421-425.Google Scholar
- Jain D, Jain S, Jain D, Amin M: Simultaneous estimation of metformin hydrochloride, pioglitazone hydrochloride, and glimepiride by RP-HPLC in tablet formulation. J Chromatogr Sci. 2008, 46: 501-504.View ArticleGoogle Scholar
- Chakradhar L, Kallem R, Karthik A, Sundari BT, Ramesh S, Mullangi R, Srinivas NR: A rapid and highly sensitive method for the determination of glimepiride in human plasma by liquid chromatography-electrospray ionization tandem mass spectrometry: application to a pre-clinical pharmacokinetic study. Biomed Chromatogr. 2008, 22: 58-63. 10.1002/bmc.896.View ArticleGoogle Scholar
- Goyal A, Singhvi I: Simultaneous spectrophotometric estimation of rosiglitazone maleate and glimepiride in tablets dosage forms. Indian Journal Pharm Sci. 2007, 69: 780-783. 10.4103/0250-474X.39433.View ArticleGoogle Scholar
- ICH Harmonized Tripartite Guideline, Validation of Analytical Procedures Text and Methodology, Q2(R1). Current Step 4 Version, Parent Guidelines on Methodology Dated November 6 1996, Incorporated in November 2005. [http://www.ich.org/fileadmin/Public_Web_Site/ICH_Products/Guidelines/Quality/Q2_R1/Step4/Q2_R1__Guideline.pdf](accessed February 15, 2008)
- Miller JN, Miller JC: Statistics and Chemometrics for Analytical Chemistry. 2005, Pearson Education Limited, Harlow, England, 256-5Google Scholar
- GlaxoSmithKline AV: L19, PRESCRIBING INFORMATION. (accessed March, 2007), [http://www.gsk-ecs.com/news/private/AvandiaAVL19P.I.pdf]
- Moffat AC, Osselton MD, Widdop B: Clark's Analysis of Drugs and Poisons in Pharmaceuticals, Body fluids and Postmortem Material. 2004, The Pharmaceutical Press, London, 1081-3Google Scholar
- Hruska MW, Frye RF: Simplified method for determination of rosiglitazone in human plasma. J Chromatogr B Analyt Technol Biomed Life Sci. 2004, 803: 317-320. 10.1016/j.jchromb.2004.01.010.View ArticleGoogle Scholar