- Open Access
Determining the chemical activity of hydrophobic organic compounds in soil using polymer coated vials
© Reichenberg et al 2008
- Received: 07 February 2008
- Accepted: 06 May 2008
- Published: 06 May 2008
In soils contaminated by hydrophobic organic compounds, the concentrations are less indicative of potential exposure and distribution than are the associated chemical activities, fugacities and freely dissolved concentrations. The latter can be measured by diffusive sampling into thin layers of polymer, as in, for example, solid phase micro-extraction. Such measurements require equilibrium partitioning of analytes into the polymer while ensuring that the sample is not depleted. We introduce the validation of these requirements based on parallel sampling into polymer layers of different thicknesses.
Equilibrium sampling devices were made by coating glass vials internally with 3–12 μm thick layers of polydimethylsiloxane (PDMS). These were filled with slurries of a polluted soil and gently agitated for 5 days. The concentrations of 7 polycyclic aromatic hydrocarbons (PAHs) in the PDMS were measured. Validation confirmed fulfilment of the equilibrium sampling requirements and high measurement precision. Finally, chemical activities of the PAHs in the soil were determined from their concentrations and activity coefficients in the PDMS.
PAHs' thermodynamic activities in a soil test material were determined via a method of uptake into PDMS. This can be used to assess chemical exposure and predict diffusion and partitioning processes.
- Partition Ratio
- Equilibrium Sampling
- Melting Point Temperature
Chemical activity and the closely related fugacity were suggested as measures of chemical behaviour and affinity by G. N. Lewis in his now century old formulation of chemical thermodynamics . Since then they have been used to understand, predict and describe physico-chemical phenomena such as osmosis , mineral stability  and cholesterol biosynthesis [4, 5]. That the chemical activity of a hydrophobic organic soil pollutant is important for its bioavailability has recently been proposed . One reason for this is that partitioning occurs spontaneously down gradients in chemical activity. Equilibrium partitioning is defined by equal chemical activities, forming the basis for equilibrium partitioning theory  and related models (e.g. [8, 9]). All of which is taken to contend that the mentioned thermodynamic quantities can help in the prediction of chemical bioaccumulation and toxicity . Despite this promising perspective we found few published methods to determine the chemical activity [11, 12] or fugacity [13–15] of hydrophobic organic pollutants in environmental samples.
In some instances chemical activities can be measured with equilibrium sampling techniques. One prominent approach, Solid Phase Micro-Extraction (SPME), was introduced in 1990 by Arthur and Pawliszyn . The SPME format is very attractive as an Equilibrium Sampling Device (ESD) because the sampling phase is a small volume (μL) of polymer with a relatively large surface area . The micro-extraction is caused by molecular diffusion across that surface, which is driven by the chemical activity of the analyte in the sample and results in a measurable concentration in the polymer. Diffusive sampling into thin layers of polymer is suitable for measurement of chemical activity and fugacity if certain conditions are fulfilled . First, equilibrium (i.e. equal chemical activity) between the sample and polymer must be reached. Second, the uptake into the polymer must not deplete the sample of analyte, as this would reduce the chemical activity in the sample . Third, polymer surface adsorption must be negligible because fugacity and activity is only proportional to the absorbed (dissolved) concentration within the polymer . This proportionality links an observable, physically real concentration, to the abstract but meaningful thermodynamic activity function. The chemical activity (a) is thus given by:
a = γP·CP
where γ denotes the activity coefficient and C the concentration of the analyte in the polymer (subscript P). At equilibrium, the analyte concentration in the polymer reveals new and important information about the sample.
Clearly, when compared to the more traditional analytical extraction techniques, equilibrium sampling differs in both means and ends. Not surprisingly they also present other challenges, and method performance must be evaluated differently. For example, Soxhlet extractions used in quantitative analytical methods are designed for complete analyte recovery, and tend to be sample destructive yielding "dirty" extracts that require further work-up before analysis. For such exhaustive extractions, it is well established that analyte recovery during extraction and clean-up can be validated by internal standards. Equilibrium sampling, in contrast, dictates that the sample must remain intact during the procedure. For example, grinding , heating and soaking soil samples in organic solvents are to be avoided as far as they destroy the matrix and might alter the chemical activity to be measured. Likewise, it must be assured that the ESD polymer is not damaged by abrasion, or its surface fouled by soil particles during the agitated equilibration procedure. These special criteria for valid equilibrium sampling are known, but inadequately addressed in practice.
Equilibrium sampling has some unusual criteria, and reliability could be gained from an integrated validation of (1st) equilibrium, (2nd) negligible depletion of sample and (3rd) negligible impact of other experimental artefacts such as adsorption on the polymer surface. This is especially important when long equilibration times (days) make separate validation experiments costly or the polymer phase is given a large surface area to volume (A/V) ratio to speed up equilibration. However, we could not find any published equilibrium sampling methods able to fulfil these validation requirements.
Diffusive samplers with multiple A/V ratios can be used to confirm equilibrium without time series measurements [17, 21]. Mechanistic arguments show that the A/V-approach allows not only validation of equilibrium, but extends to disclosing other potential artefacts such as sample depletion and polymer surface adsorption. The subsequent discussion explains the benefits of equilibrium sampling with devices of multiple coating thicknesses.
Many different ESD formats have been applied, including PDMS coated glass fibres , low-density polyethylene strips [23, 24], polyoxymethylene strips , poly(ethylene-vinyl acetate) films  and liquid-filled, hollow-fibre membranes , each with their specific operational window. Other alternatives include the Twister® (sorptive stir-bar) and the Immobilised Liquid Extraction™ Caps. Very recently Minhas et al. published work on vials internally coated with poly(ethylene-vinyl acetate) . This integration of glass container with the sampling phase allows soil and sediment samples to be transported and stored, for example, while equilibrating with the polymer coating. All the while minimising sample handling, increasing work efficiency and reducing worker exposure.
The limitations of existing ESDs and theoretical arguments suggest that equilibrium sampling with multiple coating thicknesses integrated within a sample container can provide validated and simplified measurements of chemical activity, fugacity and freely dissolved concentration. The aim of this paper is therefore to present ESDs consisting of μm-thick PDMS coatings of multiple thicknesses and their application to the measurement of chemical activity of PAHs in soil.
The ESDs consist of PDMS layers deposited on the internal wall of sample glass vials. The ESDs are equivalent in all respects apart from the thickness of the PDMS. This in vial-sample preparation minimises sample handling, and parallel application of the different thicknesses allows validation of the equilibrium sampling to measure chemical activity. The procedure consists of three parts:
1. Vials with 3–12 μm thin layers of PDMS are equilibrated with samples of the investigated soil.
2. Solvent extracts of the PDMS are analysed for the content of PAHs to determine CPDMS.
3. The chemical activity is found as the product of CPDMS and an activity coefficient (Equation 1).
Determining equilibrium partitioning concentrations in PDMS coatings
Measured equilibrium partitioning concentration in PDMS (CPDMS), chemical activity (a) and concentration (Csoil) of seven PAHs in the test soil
CPDMS (± SD) a
a n = 9-8
b dry matter
Method Limits of Quantification (LoQ)
Depletion of sample
Values of CPDMS can be applied in many different ways without any further conversion. Gradients in CPDMS follow the potential for diffusion and measured values should fall in the direction of passive transport gradients within or between environmental compartments. For example, as predictor of PCB bioaccumulation into worms, CPDMS is directly comparable between both dissimilar sediments  and soils . Likewise, CPDMS can be used to monitor environmental processes in time, such as the effect of aging or the (bio)degradation of PAHs in soil. A decrease in CPDMS is sufficient to show a decrease in chemical activity.
Determining freely dissolved concentration and chemical activity
This calibration towards Cfree has already been covered in several publications (e.g. [17, 22, 31, 32]), and the present paper focuses on the determination of chemical activities from CPDMS (Equation 1).
Activity coefficients in PDMS
Melting points (Tm), crystal activities (axstal), solubilities (S(s)MeOH) and activity coefficients (γMeOH) in methanol. Methanol-PDMSe partition ratios (KPDMS/MeOH) and activity coefficients (γPDMS) in the PDMS for 12 PAHs
a xstal a, b
S(s)MeOH (± SD) b, c
K PDMS/MeOH e
γPDMS e, f
a Equation 4
b T = 298 K (25°C)
c n = 3
d γMeOH = axstalS(s)MeOH-1
e Silastic® PDMS
f Equation 3
The practice of ESD calibration via methanol partitioning standards may prove valuable in the future. Already now, methods to conduct multi-compartment studies including equilibrium sampling measurements are available [14, 22, 23, 36]. However, these methods use different polymers and this may confound a comparison of results. For example, the inter-calibration of two commercial PDMS formulations, Silastic® and SSP-M823, revealed an almost 40% difference in their partitioning properties (Table A in Additional File 1). This kind of information is important for comparing measurements from different ESDs.
Equilibrium sampling devices with multiple A/V ratios
The in vial-ESD format and A/V-approach has unique and important features. Most important is the integrated QA/QC of the measurements. This extends the applicability towards very difficult or very dirty samples that might cause problems during equilibration or analysis due to their composition. Examples of troublesome sample components include soot particles with a high PAH content  and (micro)droplets of non-aqueous phase liquids (NAPLs) that may foul the PDMS surface. But, when using the in vial-ESD format, equilibrium sampling flaws can be inspected by plotting na against VPDMS. This integrated validation makes the method reliable at its applicability limit, whereas other equilibrium sampling methods require a greater margin of safety. Second, the integration of thin polymer layers within sample vials  reduces the risk of measurement error. Handling procedures are minimised as sample material can be secured, transported, stored and equilibrated, under seal and in a single container. The subsequent liquid extract of the PDMS makes the method compatible with many standard analytical procedures. For example, addition of internal standards, clean-up and/or pre-concentration steps can be included in future equilibrium sampling method developments. The in vial-ESDs are both practical and versatile. Third, the design of the devices is also adaptable. The ESD dimensions can be tuned towards faster equilibration by reducing the polymer coating thickness . Alternatively, detection limits can be lowered by increasing the VPDMS to sample a greater amount of analyte for detection. All of these advantages make ESDs with multiple A/V ratios attractive.
In vial-ESD preparation
3–12 μm layers of PDMS were prepared in 20 mL glass vials by dispersion coating in the following manner. The appropriate mass of medical adhesive, Silastic® PDMS Silicone Type A (Dow Corning, Seneffe, BE; density 1.15 kg/L ) paste was dispersed in 10 times its volume in n-pentane by wrist shaking and sonication. The dispersion was filtered (glass fiber, 0.7 μm nominal, Pall Corp., MI) and diluted in a geometric series. All polymer dispersions were used at the day of preparation and should be prepared using disposable glassware and materials to avoid inadvertent formation of insoluble polymer layers.
After careful tare on an analytical balance (Sartorius BP 210S, d = 0.1 mg), the vials to be coated were placed horizontally on a roller-mixer (Stuart sci., Stone, UK). While rotated (33 rpm) 3.0 mL of the appropriate dispersion was added to each vial. The solvent was evaporated under a gentle stream of nitrogen and the vials were left to polymerise for 72 h in a fume hood. Siloxane oligomers were removed from the cross-linked PDMS by three extractions with ethyl acetate (3 mL, 10 min vigorous shaking). The vials were again balanced to determine the mass of PDMS now covering their inner sidewall surface.
Equilibrium sampling procedure
PDMS-coated vials with different coating thicknesses were filled with 18–20 g of the test soil (n = 9). 10 mL aqueous solution of sodium azide (0.5 g/L) was added to each vial to inhibit microbial activity and to create a soil suspension for the sampling. Vials were capped with metal lined screw caps and placed horizontally in a customised holder rotated (6 rpm) on a roller-mixer for 122 h.
PDMS coating extraction
Soil content was removed, the vials were rinsed with a little distilled water on a whirly mixer before being wiped internally with lint-free tissue. To each vial was then added 1.00 mL methanol. The vials were again capped and rotated for 12 h. The methanol extracts were collected and stored in freezer (-20°C) until analysed for PAHs.
Methanol extracts of PDMS were analysed for PAHs by HPLC-fluorescence detection (Agilent 1100 system with G1321A FLD (Ex. 260 nm; Em. 350, 420, 440 and 500 nm). Separation column: CP-Ecospher 4 PAH (Varian Inc., Palo Alto, CA) operated at 0.5 ml/min (28°C, 10 μL injection); Mobile phase: methanol, SUPER-Q treated water (Millipore, MA); Gradient (in %methanol, by weight): t = 0 min 80%; t = 5–30 min linear gradient 80–100%; t = 30–45 min 100%. Quantification was accomplished by a five-point external standard curve. Extract analysis was carried out within 2 weeks after sampling.
Solubilities in methanol
Solubilities were measured by placing excess PAH crystals together with methanol in PTFE-sealed glass vials and equilibrating these in a thermostated water bath at 25.0 ± 0.1°C for at least 3 days. Attainment of the solubility limit was verified by repetitive measurements on consecutive days. Aliquots of the saturated methanol solutions were passed through 0.2 μm PTFE-filters into volumetric flasks and quantitatively diluted with methanol to reach suitable concentrations before analysis by the HPLC-FLD method described above.
PDMS-methanol partition ratios
Partition ratios (KPDMS/MeOH) between Silastic® PDMS polymer and methanol was measured in a separate experiment. First, PDMS sheets were prepared by filling a polyethylene mold (0.25 mm high) with Silastic® paste. The top surface was pressed even, and covered with wet tissue and a glass plate. After one week, the cross-linked PDMS was cut into sheets weighing approximately 1 g, Soxhlet extracted (ethyl acetate, 100 h) and then washed with methanol. A clean sheet and an excess of methanol solution containing all the analyte PAHs (approx. 25 μg/L) were placed in a sealed amber glass container. After equilibration by shaking for > 1 week at 20°C, the methanol and PDMS were separately taken for extraction and analysis of the PAH contents by GC-MS as described in . The respective, volume based, concentrations in the Silastic® PDMS sheet and the methanol solution were divided to calculate the partition ratios.
This study concludes that vials containing thin layers of PDMS can be used as equilibrium sampling devices. The methodology of parallel sampling in devices with different thickness PDMS layers has unique and important features. Most important is an integrated QA/QC of the procedure. When conducted on a soil test material, results were confirmed as precise and valid measurements of equilibrium partitioning concentrations, CPDMS. These were then, by way of a separate calibration experiment, expressed as chemical activities of individual PAHs in the soil.
Relevance for chemical risk assessment
A risk assessment of the pollution in the test material soil is warranted by the present soil quality criterion for PAHs in Denmark. The legislation dictates that the sum of fluoranthene, benzo [a]pyrene, benzo [k]fluoranthene, and indeno [1,2,3-cd]pyrene may not exceed 4 mg/kg dry soil .
Several concerns regarding the risk posed by the PAHs in the polluted soil can be assessed with the chemical activities reported in Table 1. First, asoil quantifies the chemical pollutant's potential for partitioning and distribution to other materials in contact with the soil. This makes the activity relevant because it is first after release and transfer to a susceptible target that a PAH may cause detriment of health or effects on the environment. Second, asoil provides a measure for comparison of chemical contamination levels. The activity reflects a chemical's affinity (e.g. partial molar free energy of sorption) to the soil material. Soils are dynamic, heterogeneous and unpredictable geosorbents. Therefore, Csoil is harder to interpret and compare than are concentrations in a well-defined, stable and homogenous phase such as PDMS. In consequence, equilibrium sampling measurements are useful because the results can easily be compared to the levels in other measured soils, in air, water or biological tissue, as well as to effect levels determined in toxicity studies, whenever such are available.
PAHs of ≥98% purity were obtained from Sigma-Aldrich (Copenhagen, DK) or Cambridge Isotope Lab. (Andover, MA).
Solvents and chemicals
P.a. grade n-pentane, ethyl acetate, sodium azide and HPLC-grade methanol were obtained from Merck (Darmstadt, DE).
The PAH contaminated soil were excavated from a former manufactured gas plant site (Gaswerk Tiefstark, Klosterwall 2, Freie Hansestadt Hamburg, DE) and then compost treated by A/S Bioteknisk Jordrens (Esbjerg, DK) prior to this study. It contained 14 % water, 5–7 % organic carbon, 11–16 % silt, 3–4 % clay and weathered coal-tar, petroleum residues, soot and ash.
The bioremediation of the soil had removed a fraction of the PAHs; whereas those associated with the black carbon  likely remained. The residual total concentration of the 16 EPA PAHs was 120 mg/kgsoil. This test material was selected on account of the PAH-containing soot that is claimed to complicate equilibrium sampling . Prior to sampling, the soil was sieved to 2 mm (steel mesh) and manually homogenized.
QA/QC and statistical treatment
The qualification criteria for PAHs extracted from the ESDs included retention times within 5% of a known sample and correct fluorescence emission wavelength(s). Signal integration was performed with HP Chemstation software (A.06.03, Agilent Tech.) and corrected by hand as necessary. About 7% standards and solvent blanks were included in the analytical sample sequences. ESD blanks were considered unnecessary since positive blank values would be eliminated during the equilibration. The measured results were tested by least squares linear regression of nPAH on VPDMS, forced through the origin. An r2 > 0.90 goodness of fit, was set as a necessary but not sufficient equilibrium sampling criterion. One outlier in the data set for anthracene was defined as the measured value of nANT differed by a factor > 2 from the predicted (n = 9). Equilibrium concentrations in PDMS (CPDMS, Table 1) were calculated as the arithmetic mean of the measurements (n = 9-8).
Activity coefficients in PDMS
where axstal is the activity of the PAH in its crystal state, S(s)MeOH the solubility in methanol and KPDMS/MeOH is the Silastic® PDMS-methanol partition ratio.
List of abbreviations
Polycyclic Aromatic Hydrocarbon
Solid Phase Micro-Extraction
Eqilibrium Sampling Device
(surface)Area to Volume ratio
Limit of Quantification
Quality Assurance/Quality Control
Chemical (thermodynamic) activity
Activity coefficient in phase i
Partition ratio between phases i and j
Concentration in (phase) i
Melting point temperature
Volume of phase i
Amount of analyte
Solubility (solid) in phase i
number of replicate determinations
Goodness of fit
Ellen Christensen (NERI, DK) measured the soil concentrations and Margit Fernquist (NERI, DK)measured solubilities. This research was funded by the Danish Technical Research Council (Young Research Leader grant), the Danish Research School of Environmental Chemistry and Ecotoxicology and the European Commission within the FP 6 Integrated Project "Novel Methods for Integrated Risk Assessment of Cumulative Stressors in the Environment" (NOMIRACLE, GOCE-003956).
- Lewis GN: Outlines of a new system of thermodynamic chemistry. Proceedings of the American Academy of Arts and Sciences. 1907, 43: 259-293.View ArticleGoogle Scholar
- Lewis GN: The osmotic pressure of concentrated solutions, and the laws of the perfect solution. Journal of the American Chemical Society. 1908, 30: 668-683. 10.1021/ja01947a002.View ArticleGoogle Scholar
- Sato M, Wright TL: Oxygen Fugacities Directly Measured in Magmatic Gases. Science. 1966, 153: 1103-1105. 10.1126/science.153.3740.1103.View ArticleGoogle Scholar
- Radhakrishnan A, Anderson TG, McConnell HM: Condensed complexes, rafts, and the chemical activity of cholesterol in membranes. Proceedings of the National Academy of Sciences of the United States of America. 2000, 97: 12422-12427. 10.1073/pnas.220418097.View ArticleGoogle Scholar
- Lee PH, Cheng DCH, Takayama K, Higuchi WI: Silicone Polymer Uptake Method for Determination of Cholesterol Thermodynamic Activity in Model Bile Systems. Journal of Pharmaceutical Sciences. 1988, 77: 610-614. 10.1002/jps.2600770712.View ArticleGoogle Scholar
- Reichenberg F, Mayer P: Two complementary sides of bioavailability: accessibility and chemical activity of organic contaminants in sediments and soils. Environmental Toxicology and Chemistry. 2006, 25: 1239-1245. 10.1897/05-458R.1.View ArticleGoogle Scholar
- Ditoro DM, Zarba CS, Hansen DJ, Berry WJ, Swartz RC, Cowan CE, Pavlou SP, Allen HE, Thomas NA, Paquin PR: Technical Basis for Establishing Sediment Quality Criteria for Nonionic Organic-Chemicals Using Equilibrium Partitioning. Environmental Toxicology and Chemistry. 1991, 10: 1541-1583. 10.1897/1552-8618(1991)10[1541:TBFESQ]2.0.CO;2.View ArticleGoogle Scholar
- Karickhoff SW: Semiempirical Estimation of Sorption of Hydrophobic Pollutants on Natural Sediments and Soils. Chemosphere. 1981, 10: 833-846. 10.1016/0045-6535(81)90083-7.View ArticleGoogle Scholar
- Mackay D, Paterson S: Calculating fugacity. Environmental Science & Technology. 1981, 15: 1006-1014. 10.1021/es00091a001.View ArticleGoogle Scholar
- Ferguson J: The use of chemical potentials as indices of toxicity. Proceedings of the Royal Society of London, Series B: Biological Sciences. 1939, 127: 387-404.View ArticleGoogle Scholar
- Legind CN, Karlson U, Burken JG, Reichenberg F, Mayer P: Determining chemical activity of (semi)volatile compounds by headspace solid-phase microextraction. Analytical Chemistry. 2007, 79: 2869-2876. 10.1021/ac061880o.View ArticleGoogle Scholar
- Ossiander L, Reichenberg F, McLachlan MS, Mayer P: Immersed Solid Phase Micro Extraction to measure chemical activity of lipophilic organic contaminants in fatty tissue samples. Chemosphere. 2008, 71: 1502-1510. 10.1016/j.chemosphere.2007.11.060.View ArticleGoogle Scholar
- Yin CQ, Hassett JP: Gas-Partitioning Approach for Laboratory and Field Studies of Mirex Fugacity in Water. Environmental Science & Technology. 1986, 20: 1213-1217. 10.1021/es00154a003.View ArticleGoogle Scholar
- Wilcockson JB, Gobas FAPC: Thin-film solid-phase extraction to measure fugacities of organic chemicals with low volatility in biological samples. Environmental Science & Technology. 2001, 35: 1425-1431. 10.1021/es001561t.View ArticleGoogle Scholar
- Golding CJ, Gobas FAPC, Birch GF: Characterization of polycyclic aromatic hydrocarbon bioavailability in estuarine sediments using thin-film extraction. Environmental Toxicology and Chemistry. 2007, 26: 829-836. 10.1897/06-378R.1.View ArticleGoogle Scholar
- Arthur CL, Pawliszyn J: Solid-phase microextraction with thermal-desorption using fused-silica optical fibers. Analytical Chemistry. 1990, 62: 2145-2148. 10.1021/ac00218a019.View ArticleGoogle Scholar
- Mayer P, Tolls J, Hermens JLM, Mackay D: Equilibrium sampling devices. Environmental Science & Technology. 2003, 37: 184A-191A. [http://pubs.acs.org/cgi-bin/article.cgi/esthag-a/0000/37/i09/pdf/503mayer.pdf]View ArticleGoogle Scholar
- Vaes WHJ, Ramos EU, Verhaar HJM, Seinen W, Hermens JLM: Measurement of the free concentration using solid-phase microextraction: Binding to protein. Analytical Chemistry. 1996, 68: 4463-4467. 10.1021/ac960337c.View ArticleGoogle Scholar
- Mayer P, Vaes WHJ, Hermens JLM: Absorption of hydrophobic compounds into the poly(dimethylsiloxane) coating of solid-phase microextraction fibers: High partition coefficients and fluorescence microscopy images. Analytical Chemistry. 2000, 72: 459-464. 10.1021/ac990948f.View ArticleGoogle Scholar
- ter Laak TL, Barendregt A, Hermens JLM: Grinding and sieving soil affects the availability of organic contaminants: A kinetic analysis. Chemosphere. 2007, 69: 613-620. 10.1016/j.chemosphere.2007.02.067.View ArticleGoogle Scholar
- Bartkow ME, Hawker DW, Kennedy KE, Muller JF: Characterizing uptake kinetics of PAHs from the air using polyethylene-based passive air samplers of multiple surface area-to-volume ratios. Environmental Science & Technology. 2004, 38: 2701-2706. 10.1021/es0348849.View ArticleGoogle Scholar
- Mayer P, Vaes WHJ, Wijnker F, Legierse K, Kraaij RH, Tolls J, Hermens JLM: Sensing dissolved sediment porewater concentrations of persistent and bioaccumulative pollutants using disposable solid-phase microextraction fibers. Environmental Science & Technology. 2000, 34: 5177-5183. 10.1021/es001179g.View ArticleGoogle Scholar
- Booij K, Hoedemaker JR, Bakker JF: Dissolved PCBs, PAHs, and HCB in pore waters and overlying waters of contaminated harbor sediments. Environmental Science & Technology. 2003, 37: 4213-4220. 10.1021/es034147c.View ArticleGoogle Scholar
- Lohmann R, Burgess RM, Cantwell GM, Ryba SA, MacFarlane JK, Gschwend PM: Dependency of polychlorinated biphenyl and polycyclic aromatic hydrocarbon bioaccumulation in mya arenaria on both water column and sediment bed chemical activities. Environmental Toxicology and Chemistry. 2004, 23: 2551-2561. 10.1897/03-400.View ArticleGoogle Scholar
- Jonker MTO, Koelmans AA: Polyoxymethylene solid phase extraction as a partitioning method for hydrophobic organic chemicals in sediment and soot. Environmental Science & Technology. 2001, 35: 3742-3748. 10.1021/es0100470.View ArticleGoogle Scholar
- Liu JF, Jonsson JA, Mayer P: Equilibrium sampling through membranes of freely dissolved chlorophenols in water samples with hollow fiber supported liquid membrane. Analytical Chemistry. 2005, 77: 4800-4809. 10.1021/ac0503512.View ArticleGoogle Scholar
- Minhas JK, Vasiluk L, Pinto LJ, Gobas F, Moore MM: Mobilization of chrysene from soil in a model digestive system. Environmental Toxicology and Chemistry. 2006, 25: 1729-1737. 10.1897/05-345R1.1.View ArticleGoogle Scholar
- Rusina TP, Smedes F, Klanova J, Booij K, Holoubek I: Polymer selection for passive sampling: A comparison of critical properties. Chemosphere. 2007, 68: 1344-1351. 10.1016/j.chemosphere.2007.01.025.View ArticleGoogle Scholar
- Trimble TA, You J, Lydy MJ: Bioavailability of PCBs from field-collected sediments: Application of Tenax extraction and matrix-SPME techniques. Chemosphere. 2008, 71: 337-344. 10.1016/j.chemosphere.2007.09.001.View ArticleGoogle Scholar
- van der Wal L, Jager T, Fleuren RHLJ, Barendregt A, Sinnige TL, Van Gestel CAM, Hermens JLM: Solid-phase microextraction to predict bioavailability and accumulation of organic micropollutants in terrestrial organisms after exposure to a field-contaminated soil. Environmental Science & Technology. 2004, 38: 4842-4848. 10.1021/es035318g.View ArticleGoogle Scholar
- Ter Laak TL, Agbo SO, Barendregt A, Hermens JLM: Freely dissolved concentrations of PAHs in soil pore water: Measurements via solid-phase extraction and consequences for soil tests. Environmental Science & Technology. 2006, 40: 1307-1313. 10.1021/es0514803.View ArticleGoogle Scholar
- Petersen DG, Reichenberg F, Dahllof I: Phototoxicity of pyrene affects benthic algae and bacteria from the Arctic. Environmental Science & Technology. 2008, 42: 1371-1376. 10.1021/es071854n.View ArticleGoogle Scholar
- Allen JO, Sarofim AF, Smith KA: Thermodynamic properties of polycyclic aromatic hydrocarbons in the subcooled liquid state. Polycyclic Aromatic Compounds. 1999, 13: 261-283. 10.1080/10406639908020550.View ArticleGoogle Scholar
- Van Noort PCM: Fugacity ratio estimations for high-melting rigid aromatic compounds. Chemosphere. 2004, 56: 7-12. 10.1016/j.chemosphere.2004.01.027.View ArticleGoogle Scholar
- Boyd BJ, Porter CJH, Charman WN: Using the polymer partitioning method to probe the thermodynamic activity of poorly water-soluble drugs solubilized in model lipid digestion products. Journal of Pharmaceutical Sciences. 2003, 92: 1262-1271. 10.1002/jps.10390.View ArticleGoogle Scholar
- Cornelissen G, Pettersen A, Broman D, Mayer P, Breedveld GD: Field Testing of Equilibrium Passive Samplers to Determine Freely Dissolved Native Polycyclic Aromatic Hydrocarbon Concentrations. Environmental Toxicology and Chemistry. 2008, 27: 499–508-10.1897/07-253.1.View ArticleGoogle Scholar
- Booij K, Smedes F, van Weerlee EM: Spiking of performance reference compounds in low density polyethylene and silicone passive water samplers. Chemosphere. 2002, 46: 1157-1161. 10.1016/S0045-6535(01)00200-4.View ArticleGoogle Scholar
- Miljøstyrelsen: Jordforureningsloven. [http://www.mst.dk/Jord/Jordforureningsloven/]
- Tsapakis M, Stephanou EG: Occurrence of gaseous and particulate polycyclic aromatic hydrocarbons in the urban atmosphere: study of sources and ambient temperature effect on the gas/particle concentration and distribution. Environmental Pollution. 2005, 133: 147-156. 10.1016/j.envpol.2004.05.012.View ArticleGoogle Scholar
- Cornelissen G, Gustafsson O, Bucheli TD, Jonker MTO, Koelmans AA, Van Noort PCM: Extensive sorption of organic compounds to black carbon, coal, and kerogen in sediments and soils: Mechanisms and consequences for distribution, bioaccumulation, and biodegradation. Environmental Science & Technology. 2005, 39: 6881-6895. 10.1021/es050191b.View ArticleGoogle Scholar
- Mackay D, Shiu WY, Ma KC: Illustrated handbook of physical-chemical properties and environmental fate for organic chemicals. Volume II: polynuclear aromatic hydrocarbons, polychlorinated dioxins and dibenzofurans. Illustrated handbook of physical-chemical properties and environmental fate for organic chemicals. 1992, Boca Raton, FL, Lewis PublishersGoogle Scholar
- Yalkowsky SH, Orr RJ, Valvani SC: Solubility and partitioning. 3. The solubility of halobenzenes in water. Industrial & Engineering Chemistry Fundamentals. 1979, 18: 351-353. 10.1021/i160072a011.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.