- Research article
- Open Access
Detection of stanozolol in environmental waters using liquid chromatography tandem mass spectrometry
© Deshmukh et al 2010
- Received: 17 June 2011
- Accepted: 14 October 2011
- Published: 14 October 2011
Owing to frequent administration of a wide range of pharmaceutical products, various environmental waters have been found to be contaminated with pharmacologically active substances. For example, stanozolol, a synthetic anabolic steroid, is frequently misused for performance enhancement as well as for illegal growth promoting purposes in veterinary practice. Previously we reported stanozolol in hair samples collected from subjects living in Budapest. For this reason we initiated this study to explore possible environmental sources of steroid contamination. The aim of this study was to develop a method to monitor stanozolol in aqueous matrices using liquid chromatography tandem mass spectrometry (LC-MS/MS).
Liquid-liquid extraction using pentane was found to be an efficient method for the extraction of stanozolol from water samples. This was followed by direct detection using LC-MS/MS. The method was capable of detecting 0.25 pg/mL stanozolol when only 5 mL water was processed in the presence of stanozolol D3 as internal standard. Fifteen bottled waters analysed were found to be negative for stanozolol. However, three out of six samples from the Danube river, collected from December '09 to November '10, were found to contain stanozolol at concentrations up to 1.82 pg/mL. In contrast, only one sample (out of six) of urban tap water from Budapest city was found to contain stanozolol, at a concentration of 1.19 pg/mL.
The method developed is efficient, rapid, reproducible, sensitive and robust for the detection of stanozolol in aqueous matrices.
- High Performance Liquid Chromatography
- Collision Induce Dissociation
- Extraction Recovery
- Selective Reaction Monitoring
- High Performance Liquid Chromatography Grade
Regular and widespread use of pharmaceuticals, which are frequently excreted as non- metabolized parent compounds, has led to growing concerns for the safety of drinking water . The vast range of pharmaceutical products that have been detected in sewage, surface, ground and drinking waters include bronchodilators, oral contraceptives, antidepressants, beta-blockers, anti-biotics, anti-inflammatories and analgesics [2–7]. Even modern sewage treatment works are not constructed to specifically eliminate pharmaceuticals  from potable water supplies.
Stanozolol, an anabolic steroid is a synthetic derivative of the endogenously-produced male-sex hormone testosterone. It is commonly misused as a performance enhancement drug because of its ability to enhance muscular strength. The World Anti-Doping Agency (WADA) has banned its use in- and out-of-competition . Despite the restriction, stanozolol is one of the most commonly misused synthetic, anabolic steroids in sport  and in veterinary practice, where it is used for growth promoting purposes .
In humans, stanozolol is mainly metabolized by undergoing hydroxylation to form mono- and di-hydroxylated metabolites. The majority of these are excreted in urine in the form of conjugates. Less than 5% are excreted as non-conjugated fractions . According to WADA, doping with stanozolol is confirmed if the urinary concentration of its major metabolite, 3-hydroxystanozolol exceeds 2 ng/mL . Unlike testosterone, the synthetic stanozolol and/or its main metabolite should only appear in environmental waters if the former is used for veterinary purposes, taken under medical supervision or illegally by athletes for performance enhancement or if either one or both of these compounds are accidently discharged into environmental waters. In previous studies, we reported the detection of stanozolol in hair samples collected from subjects living in Budapest [12, 13]. For this reason, we initiated this study to explore possible environmental sources of steroid contamination.
The aim of this study was to develop a methodology for the detection of stanozolol in aqueous matrices. To achieve this, liquid - liquid extraction (LLE) was employed for purification and concentration followed by direct determination using LC-MS/MS. Extraction recovery was evaluated for aqueous matrices spiked with stanozolol at pg/mL levels.
Reagents and chemicals
Stanozolol and stanozolol D3 (internal standard) were obtained from LGC standards (Teddington, London, UK). Pentane, deionised water, formic acid, and acetonitrile were obtained from Sigma Aldrich (Poole, Dorset, UK). All chemicals and reagents were of HPLC grade. Environmental water samples were obtained from Budapest (Hungary) and collected from the River Danube and an urban tap (drinking water) in clean, amber bottles. Water samples were collected periodically from December 2009 to November 2010. Samples from Lake Balaton and spring water (Rózsika forrás, Solymár, near Budapest) were also collected for comparison. Some commonly-consumed, bottled non-carbonated, natural mineral water samples were purchased from local supermarkets. The majority of the analysed, commercially available, bottled natural mineral waters are recognised by the European Union . All water samples were stored at -20°C and protected from light prior to analysis.
Liquid-liquid extraction (LLE) using pentane was employed for the extraction of stanozolol from water samples. Suspended particles were not filtered from the river water so that the drug adsorbed on them could also be extracted efficiently. A 5 mL aliquot of each water sample was spiked with stanozolol D3 (internal standard, 50 μL of 10 ng/mL)  followed by the addition of 3 mL pentane. The contents were vortex mixed vigorously for 10 seconds followed by centrifugation at 3500 × g at ambient temperature for 5 minutes. The pentane layer was separated and collected in a silanized glass tube. To ensure good recovery, the extraction procedure was performed twice. Both organic fractions were pooled and dried by evaporation at 45 °C under a gentle stream of nitrogen gas. The dried residue was then reconstituted with 50 μL acetonitrile. A 5 μL aliquot of the reconstituted solution was injected into the LC-MS/MS system for analysis.
LC mobile phase gradient composition
LC run time (minutes)
0.1% Formic acid in acetonitrile (%)
0.1% Formic acid in water (%)
Retention times, SRM transitions and collision energies of stanozolol and stanozolol D3 (internal standard)
Retention time (min)
Collision energy (eV)
329.2 → 81.2
329.2 → 121.2
332.2 → 81.2
The Thermo Scientific Xcalibur software (version 2.1) was used to control the LC system and mass spectrometer. Data analysis and assay performance was also evaluated using the same software. The performance of the analytical method was validated for the following set of parameters: linearity, specificity, accuracy, lower limit of detection (LLOD), lower limit of quantification (LLOQ), inter-day precision and intra-day precision. Calibration samples and quality control samples at low, medium and high concentration levels were prepared by fortifying 5 mL HPLC grade water with known concentrations of stanozolol and I.S. followed by LLE and LC-MS/MS analysis. The analyte-to-internal standard ratio was calculated by dividing the area of analyte peak by the area of the I.S. peak.
Summary of assay validation results for stanozolol
Precision RSD (%)
N = 6+6+6
N = 18+18+18
0.5 to 200
Matrix effect results for stanozolol and stanozolol D3 in HPLC water, tap water and river water
Stanozolol D3 (I.S.)
HPLC water (N = 6)
Tap water (N = 6)
River water (N = 6)
Extraction recovery of stanozolol (I.S. corrected) at 2 pg/mL
Absolute extraction recovery (%)
Relative extraction recovery (%)
HPLC water (N = 6)
Tap water (N = 6)
River water (N = 6)
Gas chromatography mass spectrometry (GC-MS) and high performance liquid chromatography (HPLC) coupled to fluorescence and UV detectors have been commonly employed to analyse steroids . However, HPLC coupled to fluorescence detection involves laborious sample preparation steps and GC-MS requires a complicated sample derivatization step which makes the method more time consuming and expensive [15, 16]. Hence, use of LC-MS/MS for analyzing steroids is a feasible approach as the sample preparation step involved is facile, economical and does not require any additional derivatization step. Compared to previous methods for detecting steroids in environmental waters [15, 17–19], the major advantage of our method is that less volume (only 5 mL, opposed to up to 1000 mL) of water sample is required for analysis. Another advantage includes the use of liquid-liquid extraction for purification of water samples, which is less time consuming and more economical in comparison to the solid phase extraction processes employed in previous studies [15, 17–20].
Water analyses results
Determination of stanozolol in environmental and domestic water samples
Environmental water sample (N = 3)
Average concentration pg/mL
31st December 2009
18th April 2010
01st September 2010
24th October 2010
05th November 2010
1.82 ± 0.19
0.71 ± 0.06
0.54 ± 0.03
1.19 ± 0.03
The possible sources of stanozolol entering the river are unknown, but may be from human or animal consumption and excretion of un-metabolized drug or due to accidental discharge of the parent compound. It should be noted that stanozolol was only found once in tap water and that this level does not present a threat to health based on recommended intake levels. Water samples from river and tap were collected periodically until November and stanozolol concentrations were found to be reducing over time as shown in Table 5. The possible reasons for a gradual reduction in concentration could be due to: i) variations in rates of contamination, ii) dilution of river water due to rise in water levels, (see Additional file 1), iii) degradation of the steroid in the river water due to other constituents in the river or photolysis, or deposition in the sediment.
The pH values of all the river and tap water samples collected were found to be in the neutral range. The pH values of bottled water analysed are summarized in Additional file 2. Stanozolol being basic in nature due to the presence of a pyrozole ring is found to be stable in neutral to slightly basic pH. Further investigation needs to be carried out for determining the source of stanozolol and reasons for gradual decrease in its concentration.
In recent years, numerous reports on steroids found in environmental waters have appeared. Stanozolol has been detected (qualitatively) in sludge samples collected from Huiyang and Meihu waste water treatment plants . Chang et al. have also reported the presence of stanozolol in Beijing influent waste water at a concentration of ca. 0.54 pg/mL . Recently, Tölgyesi et al. have reported the presence of the steroids cortisol, dexamethasone, flumethasone, prednisolone and epitestosterone in Danube river water , but their selection of analytes did not include stanozolol.
Our results indicate that stanozolol was present in the River Danube and Budapest tap water in the month of December 2009, when the water level in the river was low (Additional file 2). The National Health Service (NHS) recommends a minimum water intake of 1.2 litres every day . Hence, individuals drinking stanozolol contaminated urban tap water (1.19 pg/mL) will involuntarily consume approximately 1.43 ng stanozolol per day. Since the effective doses of stanozolol for men and women are 50-100 mgs/day and 2.5-10 mgs/day respectively , such low levels detected in drinking water may not cause significant harm to the general public, especially as they were found only at one time point. In addition, a new biological sewage treatment plant opened in July 2010 in Budapest in order to treat most of the water supplied to the city (in contrast to only 30-40% water being treated in the past). This major environmental protection investment will potentially contribute to a further decrease in levels of stanozolol compared to those we previously observed. Future studies, sampling from various river and tap water sites should, in due course, be able to provide evidence for this.
In conclusion, a rapid, highly sensitive, robust and reproducible method has been developed to detect stanozolol in different types of water samples. The assay is capable of detecting stanozolol at a concentration as low as 0.25 pg/mL water when only 5 mL water is processed. The performance of this method gives acceptable relative recoveries for stanozolol river and tap water samples. The method can be extended to detect other chemicals and pharmaceutical drugs which may be hazardous to human health and environment.
The authors thank Julian Swinden and Iltaf Shah for technical advice and help; IP and FP for sample collection. The photos were taken by FP and ZsB.
- Rodriguez-Mozaz S, Weinberg S: Meeting report: pharmaceuticals in water--An interdisciplinary approach to a public health challenge. Environ Health Perspect. 2010, 118 (7): 1010-1020. 10.1289/ehp.0901648.View ArticleGoogle Scholar
- Costanzo S, Murby J, Bates J: Ecosystem response to antibiotics entering the aquatic environment. Mar Pollut Bull. 2005, 51: 218-223. 10.1016/j.marpolbul.2004.10.038.View ArticleGoogle Scholar
- Ternes TA: Occurrence of drugs in German sewage treatment plants and rivers. Water Res. 1998, 32: 3245-3268. 10.1016/S0043-1354(98)00099-2.View ArticleGoogle Scholar
- Ye Z, Weinberg HS, Meyer MT: Trace analysis of trimethoprim and sulfonamide, macrolide, quinolone, and tetracycline antibiotics in chlorinated drinking water using liquid chromatography electrospray tandem mass spectrometry. Anal Chem. 2007, 79: 1135-1144. 10.1021/ac060972a.View ArticleGoogle Scholar
- Heberer T: Occurrence, fate, and removal of pharmaceutical residues in the aquatic environment: a review of recent research data. Toxicol Lett. 2002, 131: 5-17. 10.1016/S0378-4274(02)00041-3.View ArticleGoogle Scholar
- Glassmeyer S, Koplin D, Furlong E, Focazio M: Environmental presence and persistence of pharmaceuticals: an overview. Fate of Pharmaceuticals in the Environment and in Water Treatment Systems. Edited by: Aga DS. 2008, Boca Raton, FL: CRC Press, Taylor and Francis Books, 3-51.Google Scholar
- Daughton CG, Ternes TA: Pharmaceuticals and personal care products in the environment: agents of subtle change?. Environ Health Perspect. 1999, 107 (6): 907-938. 10.1289/ehp.99107s6907.View ArticleGoogle Scholar
- World Anti-Doping Agency: The 2011 Prohibited List. [accessed on 11.06.2011], [http://www.wada-ama.org/Documents/World_Anti-Doping_Program/WADP-Prohibited-list/To_be_effective/WADA_Prohibited_List_2011_EN.pdf]
- World Anti-Doping Agency: Laboratory statistics. [accessed on 09.06.2011], [http://www.wada-ama.org/rtecontent/document/WADA_2008_LaboratoryStatisticsReport_Final.pdf]
- Poelmans S, Wasch KD, De Brabander HF, Van De Wiele M, Courtheyn D: Analytical possibilities for the detection of stanozolol and its metabolites. Anal Chim Acta. 2002, 473: 39-47. 10.1016/S0003-2670(02)00672-4.View ArticleGoogle Scholar
- World Anti-Doping Agency: Technical Document - TD2010MRPL. [accessed on 10.06.2011], [http://www.wada-ama.org/Documents/World_Anti-Doping_Program/WADP-IS-Laboratories/WADA_TD2010MRPLv1.0_Minimum%20Required%20Performance%20Levels_Sept%2001%202010_EN.pdf]
- Petróczi A, Aidman EV, Hussain I, Deshmukh N, Nepusz T, Uvacsek M, Tóth M, Barker J, Naughton DP: Virtue or Pretense? Looking behind Self-Declared Innocence in Doping. PLoS One. 2010, 5 (5): 11-View ArticleGoogle Scholar
- Deshmukh N, Hussain I, Barker J, Petroczi A, Naughton DP: Analysis of anabolic steroids in human hair using LC-MS/MS. Steroids. 2010, 75 (10): 710-714. 10.1016/j.steroids.2010.04.007.View ArticleGoogle Scholar
- List of natural mineral waters recognised by member states. [accessed on 09.06.2011], [http://ec.europa.eu/food/food/labellingnutrition/water/mw_eulist_en.pdf]
- Tölgyesi A, Verebey Z, Sharma VK, Kovacsics L, Fekete J: Simultaneous determination of corticosteroids, androgens, and progesterone in river water by liquid chromatography-tandem mass spectrometry. Chemosphere. 2010, 78: 972-979. 10.1016/j.chemosphere.2009.12.025.View ArticleGoogle Scholar
- Kushnir MM, Rockwood AL, Roberts WL, Yue B, Bergquist J, Meikle AW: Liquid chromatography tandem mass spectrometry for analysis of steroids in clinical laboratories. Clin Biochem. 2011, 45 (1): 77-88.View ArticleGoogle Scholar
- Liu S, Ying GG, Zhao JL, Chen F, Yang B, Zhou LJ, Lai HJ: Trace analysis of 28 steroids in surface water, wastewater and sludge samples by rapid resolution liquid chromatography-electrospray ionization tandem mass spectrometry. J Chromatogr A. 2011, 1218 (10): 1367-1378. 10.1016/j.chroma.2011.01.014.View ArticleGoogle Scholar
- Chang H, Wan Y, Wu S, Fan Z, Hu J: Occurrence of androgens and progestogens in wastewater treatment plants and receiving river waters: Comparison to estrogens. Water Res. 2011, 45 (2): 732-740. 10.1016/j.watres.2010.08.046.View ArticleGoogle Scholar
- Chang H, Wu S, Hu J, Asami M, Kunikane S: Trace analysis of androgens and progestogens in environmental waters by ultra-performance liquid chromatography-electrospray tandem mass spectrometry. J Chromatogr A. 2008, 1195: 44-51. 10.1016/j.chroma.2008.04.055.View ArticleGoogle Scholar
- Chang H, Wan Y, Hu J: Determination and source apportionment of five classes of steroid hormones in urban rivers. Environ Sci Technol. 2009, 43: 7691-7698. 10.1021/es803653j.View ArticleGoogle Scholar
- NHS: A balanced diet. [accessed on 09.06.2011], [http://www.nhs.uk/Conditions/Diet/Pages/Fluids.aspx]
- Steroid: Winstrol. [accessed on 09.06.2011], [http://www.steroid.com/Winstrol.php]